The CH SELECT switch has four bank settings, which assign EBI100C output (i.e., Magnitude or Phase) channels as follows:
|Bank||Magnitude (MAG)||Phase (PHS)|
|1||Channel 1||Channel 9|
|2||Channel 2||Channel 10|
|3||Channel 3||Channel 11|
|4||Channel 4||Channel 12|
If a particular EBI100C output is not used, the respective assigned channel cannot be used for another modules output; simply do not record on the assigned but unwanted channel.
Possible Lead Configurations
Methodological Guidelines for Impedance Cardiography , PSYCHOPHYSIOLOGY, Vol. 27, No. 1
See more on the Support tab > Downloads/Resources.