The CH SELECT switch has four bank settings, which assign EBI100C output (i.e., Magnitude or Phase) channels as follows:
|Bank||Magnitude (MAG)||Phase (PHS)|
|1||Channel 1||Channel 9|
|2||Channel 2||Channel 10|
|3||Channel 3||Channel 11|
|4||Channel 4||Channel 12|
If a particular EBI100C output is not used, the respective assigned channel cannot be used for another modules output; simply do not record on the assigned but unwanted channel.
Possible Lead Configurations
Methodological Guidelines for Impedance Cardiography , PSYCHOPHYSIOLOGY, Vol. 27, No. 1
Yadollahi, Azadeh & Singh, B & Bradley, T. (2015). Investigating the Dynamics of Supine Fluid Redistribution Within Multiple Body Segments Between Men and Women. Annals of biomedical engineering. 43. 10.1007/s10439-015-1264-0.
- This study used four EBI100C impedance measurement modules to measure fluid volume in various body segments; for each segment, two electrodes injected high frequency ([12.5–100] kHz), low amplitude (400 lA) current to the segment and two electrodes measured voltage to estimate bioelectrical impedance.
See more on the Support tab > Downloads/Resources.