If a BioNomadix wireless physiology signal displays dropout/flatline…
On a single channel
Make sure transducer/lead is firmly inserted to transmitter channel, electrodes are fully adhered, and leads are not pulling
Switch recording channels (both X, Y, and various channels on Receiver)
On both channels
Radio interference is likely disrupting transmission. Try these steps to restore signal quality:
Locate the Receiver/Logger/Smart Center closer to the Transmitter
Charge the Transmitter
Eliminate alternate sources of electrical noise (i.e., cell phones)
Elevate the BioNomadix Receiver/Logger/Smart Center higher in the room
Maintain line of sight to the Receiver/Logger/Smart Center
Participants may inadvertently be blocking line of sight when they move (i.e., scrunch their head to their shoulder). If there’s enough length in the leads, mount Transmitter on the shoulder or upper arm to let more of the radio energy escape the Transmitter.
If the recording environment is near a military base, it’s possible that military-grade RF equipment is interfering; a spectral scan would be required to confirm that and then frequencies could be positioned at quiet points (via BIOPAC Support Request).
Current shipping Firmware can’t be used to change to fixed frequencies on paired BioNomadix Receivers; Loggers and Smart Centers can specify their channels.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]
Physiological Lung Testing in the MRI to Find Damage from Military Deployments Our military veterans who served overseas in the middle east often come home with scars both seen and unseen. For some vets, unfortunately, these scars can take the form of lung damage sustained due to sandstorms, burn pits, and other high toxicity situations apparent […]