If a BioNomadix wireless physiology signal displays dropout/flatline…
On a single channel
Make sure transducer/lead is firmly inserted to transmitter channel, electrodes are fully adhered, and leads are not pulling
Switch recording channels (both X, Y, and various channels on Receiver)
On both channels
Radio interference is likely disrupting transmission. Try these steps to restore signal quality:
Locate the Receiver/Logger/Smart Center closer to the Transmitter
Charge the Transmitter
Eliminate alternate sources of electrical noise (i.e., cell phones)
Elevate the BioNomadix Receiver/Logger/Smart Center higher in the room
Maintain line of sight to the Receiver/Logger/Smart Center
Participants may inadvertently be blocking line of sight when they move (i.e., scrunch their head to their shoulder). If there’s enough length in the leads, mount Transmitter on the shoulder or upper arm to let more of the radio energy escape the Transmitter.
If the recording environment is near a military base, it’s possible that military-grade RF equipment is interfering; a spectral scan would be required to confirm that and then frequencies could be positioned at quiet points (via BIOPAC Support Request).
Current shipping Firmware can’t be used to change to fixed frequencies on paired BioNomadix Receivers; Loggers and Smart Centers can specify their channels.
fNIRS Imaging Systems are stand-alone functional brain imaging solutions for continuous NIR spectroscopy (NIRS). They provide an in-lab cognitive function assessment and eliminate a great many of the drawbacks of a functional MRI. fNIRS optical imaging technology measures neural activity and hemodynamic response in the prefrontal cortex. Subjects wear an fNIR sensor (IR light sources […]
BIOPAC provides software and hardware that allows research teams to study dyadic interactions. Here are a few notable studies using BIOPAC equipment for dyadic research focusing on ECG and EDA data. Are Frustrated Kids Leading to Frustrated Parents? Ask anyone and they will say there is no one definitive way to parent a child. Any parent can […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]