If a BioNomadix wireless physiology signal displays dropout/flatline…
On a single channel
Make sure transducer/lead is firmly inserted to transmitter channel, electrodes are fully adhered, and leads are not pulling
Switch recording channels (both X, Y, and various channels on Receiver)
On both channels
Radio interference is likely disrupting transmission. Try these steps to restore signal quality:
Locate the Receiver/Logger/Smart Center closer to the Transmitter
Charge the Transmitter
Eliminate alternate sources of electrical noise (i.e., cell phones)
Elevate the BioNomadix Receiver/Logger/Smart Center higher in the room
Maintain line of sight to the Receiver/Logger/Smart Center
Participants may inadvertently be blocking line of sight when they move (i.e., scrunch their head to their shoulder). If there’s enough length in the leads, mount Transmitter on the shoulder or upper arm to let more of the radio energy escape the Transmitter.
If the recording environment is near a military base, it’s possible that military-grade RF equipment is interfering; a spectral scan would be required to confirm that and then frequencies could be positioned at quiet points (via BIOPAC Support Request).
Current shipping Firmware can’t be used to change to fixed frequencies on paired BioNomadix Receivers; Loggers and Smart Centers can specify their channels.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]