If a BioNomadix wireless physiology signal displays dropout/flatline…
On a single channel
Make sure transducer/lead is firmly inserted to transmitter channel, electrodes are fully adhered, and leads are not pulling
Switch recording channels (both X, Y, and various channels on Receiver)
On both channels
Radio interference is likely disrupting transmission. Try these steps to restore signal quality:
Locate the Receiver/Logger/Smart Center closer to the Transmitter
Charge the Transmitter
Eliminate alternate sources of electrical noise (i.e., cell phones)
Elevate the BioNomadix Receiver/Logger/Smart Center higher in the room
Maintain line of sight to the Receiver/Logger/Smart Center
Participants may inadvertently be blocking line of sight when they move (i.e., scrunch their head to their shoulder). If there’s enough length in the leads, mount Transmitter on the shoulder or upper arm to let more of the radio energy escape the Transmitter.
If the recording environment is near a military base, it’s possible that military-grade RF equipment is interfering; a spectral scan would be required to confirm that and then frequencies could be positioned at quiet points (via BIOPAC Support Request).
Current shipping Firmware can’t be used to change to fixed frequencies on paired BioNomadix Receivers; Loggers and Smart Centers can specify their channels.
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
A System to Measure Physiological Response During Social Interaction in VR Children with Autism Spectrum Disorder (ASD) have a difficult experience in responding to social cues and actions. This pilot study measured the effectiveness of Virtual Reality in creating a social environment for children, measured with a BIOPAC MP Series data acquisition system and Worldviz […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]