The UIM100C is bidirectional. With a BIOPAC amplifier connected to the UIM100C, the appropriate input becomes an output.
For example, if you have an ECG100C module set to channel 1, you can connect a cable to the channel 1 connector on the front of the UIM100C to access the analog signal. Use the CBL100 series connection cables to interface with the UIM100C.
Please note: To maintain subject electrical isolation within the MP system and when interfacing BIOPAC hardware with other third-party products that are mains powered, use the HLT100C module in conjunction with the OUTISO or INISO.
Use the OUTISO to connect MP analog signal outputs (amplifier and data acquisition unit) to external equipment inputs.
Use the INISO to connect external equipment outputs to MP analog input channels.
Each signal isolator comes with an RJ11 cable for connection to the HLT100C module. These adapters are required to maintain subject isolation when interfacing BIOPAC hardware with other third-party products.
Alex Dimov, BIOPAC Sales Executive, was interviewed at the Neuromarketing World Forum by Inside Marketing to discuss how virtual reality can improve marketing research. Alex explains how VR can be used to present virtual iterations of a product to save time and money, and notes that measuring physiological responses adds lots of information for product and messaging […]
February 14, 2019 BIOPAC Systems, Inc. Goleta, CA USA New “Smart Amplifiers” make it easier than ever for life science researchers to get great data Fast Setup & High Performance Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level voltage connection to the data acquisition system and reduces noise artifact. […]