The UIM100C is bidirectional. With a BIOPAC amplifier connected to the UIM100C, the appropriate input becomes an output.
For example, if you have an ECG100C module set to channel 1, you can connect a cable to the channel 1 connector on the front of the UIM100C to access the analog signal. Use the CBL100 series connection cables to interface with the UIM100C.
Please note: To maintain subject electrical isolation within the MP system and when interfacing BIOPAC hardware with other third-party products that are mains powered, use the HLT100C module in conjunction with the OUTISO or INISO.
Use the OUTISO to connect MP analog signal outputs (amplifier and data acquisition unit) to external equipment inputs.
Use the INISO to connect external equipment outputs to MP analog input channels.
Each signal isolator comes with an RJ11 cable for connection to the HLT100C module. These adapters are required to maintain subject isolation when interfacing BIOPAC hardware with other third-party products.
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
A System to Measure Physiological Response During Social Interaction in VR Children with Autism Spectrum Disorder (ASD) have a difficult experience in responding to social cues and actions. This pilot study measured the effectiveness of Virtual Reality in creating a social environment for children, measured with a BIOPAC MP Series data acquisition system and Worldviz […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]