The UIM100C is bidirectional. With a BIOPAC amplifier connected to the UIM100C, the appropriate input becomes an output.
For example, if you have an ECG100C module set to channel 1, you can connect a cable to the channel 1 connector on the front of the UIM100C to access the analog signal. Use the CBL100 series connection cables to interface with the UIM100C.
Please note: To maintain subject electrical isolation within the MP system and when interfacing BIOPAC hardware with other third-party products that are mains powered, use the HLT100C module in conjunction with the OUTISO or INISO.
Use the OUTISO to connect MP analog signal outputs (amplifier and data acquisition unit) to external equipment inputs.
Use the INISO to connect external equipment outputs to MP analog input channels.
Each signal isolator comes with an RJ11 cable for connection to the HLT100C module. These adapters are required to maintain subject isolation when interfacing BIOPAC hardware with other third-party products.
The ETVision (ETV) wearable unit resembles an eyeglasses frame that can be worn by itself or over the participant’s prescription eyeglasses. It contains miniature “eye cameras” that view each eye and a “scene camera” that views the scene in front of the participant, and a microphone. An optional visor can be mounted to the front […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]