The UIM100C is bidirectional. With a BIOPAC amplifier connected to the UIM100C, the appropriate input becomes an output.
For example, if you have an ECG100C module set to channel 1, you can connect a cable to the channel 1 connector on the front of the UIM100C to access the analog signal. Use the CBL100 series connection cables to interface with the UIM100C.
Please note: To maintain subject electrical isolation within the MP system and when interfacing BIOPAC hardware with other third-party products that are mains powered, use the HLT100C module in conjunction with the OUTISO or INISO.
Use the OUTISO to connect MP analog signal outputs (amplifier and data acquisition unit) to external equipment inputs.
Use the INISO to connect external equipment outputs to MP analog input channels.
Each signal isolator comes with an RJ11 cable for connection to the HLT100C module. These adapters are required to maintain subject isolation when interfacing BIOPAC hardware with other third-party products.
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
BIOPAC provides software and hardware that allows research teams to record and analyze numerous body signals for physiological experimentation. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and Respiration data. What Keeps Students More Engaged? The question has been brought to the forefront during COVID, what variations of online learning keep students more engaged? […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]