The UIM100C is bidirectional. With a BIOPAC amplifier connected to the UIM100C, the appropriate input becomes an output.
For example, if you have an ECG100C module set to channel 1, you can connect a cable to the channel 1 connector on the front of the UIM100C to access the analog signal. Use the CBL100 series connection cables to interface with the UIM100C.
Please note: To maintain subject electrical isolation within the MP system and when interfacing BIOPAC hardware with other third-party products that are mains powered, use the HLT100C module in conjunction with the OUTISO or INISO.
Use the OUTISO to connect MP analog signal outputs (amplifier and data acquisition unit) to external equipment inputs.
Use the INISO to connect external equipment outputs to MP analog input channels.
Each signal isolator comes with an RJ11 cable for connection to the HLT100C module. These adapters are required to maintain subject isolation when interfacing BIOPAC hardware with other third-party products.
MP160 data acquisition and analysis systems with AcqKnowledge software provide a flexible tool for life science research. MP160 data acquisition workstations include all hardware, software, cables, and guides you need to get started. MP160 Data Acquisition Hardware—Ethernet-ready data acquisition and analysis unit can record multiple channels with differing sample rates; record at speeds up to 400 kHz (aggregate). […]
BIOPAC provides software and hardware that allows engineers to study the body. Here are a few notable studies using BIOPAC equipment for Biomedical Engineering research. Rehabilitating the Rehab Industry Could physical therapy be done from the comforts of your own home? If possible, this would open the door for more people with all sorts of ailments to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]