MP150: Snap the UIM100C onto the right side of the MP150.
MP100: Connect the UIM100C to the MP100 acquisition unit via two included 0.6-meter cables:
o 37-pin connector for analog signals
o 25-pin connector for digital signals
When using the Universal Interface Module (UIM100C) with other 100-Series modules, the UIM100C is usually the first module cascaded in the chain. If using the STM100C, OXY100C or HLT100C, the module must be plugged in on the left of the UIM100C. Up to seventeen modules (including the UIM100C) can be snapped together.
See Cables & Accessories for cables to interface your equipment to the UIM100C.
IMPORTANT USAGE NOTE
Mains powered external laboratory equipment should not be connected to an MP150 or MP100 System (via the UIM100C), when the system also connects to electrodes attached to humans. For analog signals, use the INISO or OUTISO (using the HLT100C) to couple external equipment to an MP System. For digital signals, use the STP100 (using the UIM100C) to couple external equipment to an MP150 or MP100 System. Contact BIOPAC for details.