Make sure amplifiers are not set to the same channel—even if some of the amplifiers are not used.
Check the switch on top of each amplifier (for some amplifiers the switch is in the front because they use multiple channels).
Calibrate and scale the signals (MP Menu > Set Up Channels).
Set the sample rate(s) and recording length (MP Menu > Set Up Acquisition).
Adjust the display scales so the data is visible (Display menu > Autoscale waveforms).
Use correct ground and avoid ground loops.
To record EDA/GSR with other biopotential signals (ECG, EEG, EOG, EGG, EMG, ERS), BIOPAC recommends using CBL205 connected to one ground on any of the biopotential amplifiers. The subject will be grounded through the Vin- of the EDA electrodes, but in some cases it is necessary to have more than one ground; in such cases, use an AC-coupled lead adapter (CBL205) to prevent ground loops.
For example, if, while recording a biopotential and EDA/GSR you have to remove the EDA electrode during a stage of the experiment, you will still want to have a ground for the biopotential. Connect the Vin- lead of the EDA as ground, and connect an AC-coupled ground to the biopotential amplifier. The result is that you will always have a ground and no ground loops.
Ground loops can be a problem when using any two of the following amplifiers at the same time: EDA100C (or GSR100C), EBI100C, or NICO100C, but there are setup options if absolute measurements (i.e., SCL) are required:
Use two wired amplifiers (EDA100C/GSR100C with NICO100C) with a single MP150 System by adding the Isolated power supply IPS100C, interface amplifier HLT100C, isolator OUTISO, and cable CBL100.
Use a single MP150 system and opt to record EDA/GSR wirelessly and Cardiac Output wired.
Use two MP150 systems, with one of the two units connected to each system or two BioNomadix amplifiers.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
Studying Cognitive Activity with fNIR! The age-old game of chess is known as a thinking man’s game, where foresight, critical thinking, and strategy are required to become a good chess player, and the terms “genius” or “brilliance” are bestowed upon the chess masters of the world. As such, chess is an excellent cognitive tool to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]