It is possible to validate the signals coming from any BIOPAC biopotential amplifier* by using a CBLCALC Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ±10 V range.
Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCALC is connected to. The CBLCALC will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different from the source signal level, contact BIOPAC for further assistance.
Smart Center Stand-alone, Portable Wireless Physiology Systems Smart Center Essentials entry-level system includes a Smart Center Device (BN-SMART with USB power cable), 2 or 3 Transmitters (BN-xxx-T with chargers); AcqKnowledge for Smart Center, and a case. Smart Center Enhanced adds a Logger (BN-LOGGER) and Basic Scripting License (ACK100W-BAS) to the Smart Center Device (BN-SMART with powercable), 3 […]
BIOPAC offers complete systems for life science research and education—human, animal, or in vivo studies, as well as powerful software and automated analysis including industry-standard data acquisition amplifiers, transducers & electrodes. BIOPAC can be used inside or outside the lab to measure physiology anytime, anyplace. Impact of Movie Editing on Perceptual Conflict Mechanisms Our ability to watch and understand films effortlessly […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]