It is possible to validate the signals coming from any BIOPAC biopotential amplifier* by using a CBLCALC Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ±10 V range.
Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCALC is connected to. The CBLCALC will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different from the source signal level, contact BIOPAC for further assistance.
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
BIOPAC provides software and hardware that allows engineers to study the body. Here are a few notable studies using BIOPAC equipment for Biomedical Engineering research. Rehabilitating the Rehab Industry Could physical therapy be done from the comforts of your own home? If possible, this would open the door for more people with all sorts of ailments to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]