It is possible to validate the signals coming from any BIOPAC biopotential amplifier* by using a CBLCALC Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ±10 V range.
Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCALC is connected to. The CBLCALC will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different from the source signal level, contact BIOPAC for further assistance.
Basic principles of physiological data collection covers fundamentals of data collection and equipment configuration equipment to help you refine experiment protocols and avoid costly missteps. Frazer Findlay, CEO of BIOPAC, discusses common mistakes with physiology recording and shares his secrets for collecting great data. Topics include: How to prep a subject and where to place electrodes; Which type of electrodes work for different body signals; Analog to Digital Conversion; Sample rates (correct vs. incorrect); Filtering; and Scaling and calibration.Watch On Demand Now!
Recent Exercise Physiology studies feature BIOPAC’s MP Research series of Data Collection recorders and AcqKnowledge software. These studies promote healthy living and observe muscle response to normal physiological motions during exercise! Trampoline Workouts for Seniors? As we get older, our joints ache and muscles stiffen—but scientists are finding new ways to not only help us […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]