It is possible to validate the signals coming from any BIOPAC biopotential amplifier* by using a CBLCALC Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ±10 V range.
Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCALC is connected to. The CBLCALC will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different from the source signal level, contact BIOPAC for further assistance.
Smart shirt simultaneously acquires ECG and Respiration data while subjects roam freely The lightweight, comfortable BioNomadix BioShirt contains a respiration sensor and fabric electrodes to wirelessly record both respiration and ECG while ambulatory subjects move freely and perform tasks in short or long-term studies, in the lab, or in the real world. The shirt connects to a wireless BioNomadix BN-RSPEC Respiration […]
Research posters presented at BIOPAC’s T4 physiology conference (Tools, Trends, Techniques, and Technology) exhibited leading physiological research utilizing BIOPAC’s technology. Research posters presented unique advances in neuromarketing and health research. The T4 conference, held at UCSB Monday, July 24, 2017 through Wednesday, July 26, 2017, showcased the latest trends in the Tech industry’s physiological research. […]
ABC News | Good Morning America | Excessive Cell Phone Use JULY 31, 2017 | Excessive cell phone use may cause anxiety, experts warn BIOPAC user Nancy A. Cheever, Ph.D. from California State University, Dominguez Hills was featured on Good Morning America. Her study investigated whether spending too much time on your phone may be causing people to feel […]