CBLCAL is a device used to check the accuracy of BIOPAC biopotential amplifiers. CBLCAL stands for cable calibration and is used to calibrate EMG, EOG, EEG, ECG, and ERS amplifiers. It functions as an adapter that allows the MP100 System to acquire a calibrated signal (through a biopotential amplifier) which is output simultaneously by the MP100 System. Using a known stimulus signal, output via the UIM100A, the CBLCAL attenuates this signal (divides it by 1,000) and re-routes it back into the amplifier where it is amplified by the selected gain setting (for instance, 1,000).
Accordingly, a one-volt input signal maps to a one-volt signal acquired in the acquisition window. This information can then be used to check for proper calibration of the amplifier.
EEG: Electroencephalography - Wired and wireless EEG solutions. Record up to 32 channels of EEG and use software features for filtering, removing EOG artifacts, and complete frequency analysis.
ERS: Evoked Response - Record evoked, late, and field potentials, as well as startle and nerve conduction. Powerful on- and off-line averaging features (P300), and dual stimulation ability with the MP150.
Psychophysiology - Record and analyze BP, ECG, HRV, EDA, EMG, EEG, EOG, RSP, etc. Interface to stimulus presentation programs...use automated analysis routines to easily score and analyze data.
EOG: Eye Movement - Record eye position and movement with the EOG100C amplifier. Find frequency, distance, velocity, and point of focus. Isolate behavior associated with nystagmus and saccades.
Sleep Studies - Long term recordings with up to 16 channels of data. Record EEG, EOG, EMG, respiration, temp., sound, limb position and more. Filter out EEG frequencies to score sleep stages.
ECG: Cardiology - Connect up to 16 ECG input leads with MP160 system. Use automated analysis features for HRV, RSA, to classify heartbeats, identify arrhythmias, and perform ECG averaging.
Cardiovascular Hemodynamics - Hardware for human and animals. Measure continuous BP, ECG, stroke volume, cardiac output, PPG, etc. For animal and tissue look at acute and chronic dose-response.
EGG: Electrogastrogram - Measure electrical activity of gastrointestinal system in humans and animals. Examine slow wave propagation and perform automated gastric wave analysis in AcqKnowledge.
Exercise Physiology - Examine ventilation, oxygen uptake, carbon dioxide production, biopotentials, temp., and biomechanical signals simultaneously. Record wirelessly with BioNomadix.
EMG: Electromyography - Record surface, needle, and fine wire EMG—or wireless EMG with BioNomadix. Use real-time integration tools for immediate analysis or use automated EMG analysis routines.
Biomechanics - Use BIOPAC equipment for gait analysis, goniometry, ergonomics evaluation, range of motion, EMG, Remote Monitoring or to interface with third-party equipment.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
BIOPAC provides software and hardware that allows research teams to study dyadic interactions. Here are a few notable studies using BIOPAC equipment for dyadic research focusing on ECG and EDA data. Are Frustrated Kids Leading to Frustrated Parents? Ask anyone and they will say there is no one definitive way to parent a child. Any parent can […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]