VREF, excitation voltage and reference voltage refer to the same concept. This is the voltage the DA100C series amplifier provides for connected transducers that need to be powered. The excitation voltage level (peak-to-peak, i.e. the absolute difference between the value for VREF- and VREF+) will influence the readings obtained from the transducer. Therefore, this voltage must be known for the purposes of calibration. The Reference Calibrator for DA100C (REFCAL) can be used. The REFCAL plugs into the DA100C and directs the reference voltage setting to the DA100C output. The REFCAL makes it very easy to adjust the reference voltage of the DA100C to suit your transducer. Alternatively, a voltmeter can be used.
Cardiovascular insights are much easier to achieve with this new integration from BIOPAC and CNSystems. The CNAP™ technology is now fully incorporated in a single BIOPAC amplifier for the MP160 system where it can be fully controlled by AcqKnowledge Software—optimized for research utilizing core CNAP clinical technology.
What you will learn: • Correct equipment set up and operation • Principles of the CNAP measurement technology • How to get reproducible data • How to interface equipment and optimize multiple measurements • Automatic blood pressure analysis • Application of the “Focus Areas” feature in AcqKnowledge Software
BIOPAC provides software and hardware that allows research teams to study physiological measures related to stress. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and EEG/ERP. Moms Moderating Stress Can relaxation techniques help lower stress on both mother and fetus? Using an MP36R Data Acquisition system to record ECG and EDA, this study […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]