VREF, excitation voltage and reference voltage refer to the same concept. This is the voltage the DA100C series amplifier provides for connected transducers that need to be powered. The excitation voltage level (peak-to-peak, i.e. the absolute difference between the value for VREF- and VREF+) will influence the readings obtained from the transducer. Therefore, this voltage must be known for the purposes of calibration. The Reference Calibrator for DA100C (REFCAL) can be used. The REFCAL plugs into the DA100C and directs the reference voltage setting to the DA100C output. The REFCAL makes it very easy to adjust the reference voltage of the DA100C to suit your transducer. Alternatively, a voltmeter can be used.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
MIT and Harvard researchers developed a new technique to expand ventilator capacity Individualized System for Augmenting Ventilator Efficacy (iSAVE): A Rapidly deployable system to expand ventilator capacity “The iSAVE is a rapidly deployable solution in response to the urgent shortage of mechanical ventilators for respiratory support during the COVID-19 pandemic. In contrast to previously described […]
New Ventilator Validation System Allows Device Manufacturers to Test Ventilator Pressure and Volume System supports rapid development and testing of hardware for COVID-19 treatment Goleta, California – March 24, 2020 BIOPAC Systems, Inc. announced a new Ventilator Validation System for engineering and device manufacturers to use when designing and testing new and existing ventilators for […]