VREF, excitation voltage and reference voltage refer to the same concept. This is the voltage the DA100C series amplifier provides for connected transducers that need to be powered. The excitation voltage level (peak-to-peak, i.e. the absolute difference between the value for VREF- and VREF+) will influence the readings obtained from the transducer. Therefore, this voltage must be known for the purposes of calibration. The Reference Calibrator for DA100C (REFCAL) can be used. The REFCAL plugs into the DA100C and directs the reference voltage setting to the DA100C output. The REFCAL makes it very easy to adjust the reference voltage of the DA100C to suit your transducer. Alternatively, a voltmeter can be used.
New series of unshielded 45 cm electrode leads with thin cable; suitable for facial EMG and other areas where lighter, shorter lead cables are required. Use for female Touchproof connectors to pinch clip connectors—connect electrodes to either a C-series amplifier or an MEC. The pinch connectors are light weight and the lead cable tinsel wire […]
Academics and professionals in the field of human physiology gathered at UCSB for the inaugural T4 Physiology Conference. BIOPAC coordinated the three day conference held in late July, which attracted attendees from 26 different countries. The attendees gathered to learn the latest tools, trends, techniques and technology used in life science measurement and interpretation. The […]
Industry reporting firm medGadget recognized BIOPAC as a major market competitor in the global Blood Flow Measurement Devices market. BIOPAC hardware solutions for blood flow measurement include laser Doppler flowmetry and extensive online and offline analysis of raw blood flow data to take an array of calculations and measurements, such as Pulsatile max, Pulsatile Min, […]