When the amplified signal for any given channel exceeds the range -10 to +10 volts, the signal will rail. You will see a straight line at -10 or + 10 volts (more likely the reading will be close 9.99 volts). The MP system is designed to work only in the range -10 to +10 volts. The signal could rail for several reasons (which are not exclusive):
The signal is too strong for the current gain setting on the amplifier. Try reducing the gain setting.
You are recording in DC mode when you should be recording in AC mode. If this is the case, make sure DC is off and a switch is set to 0.05Hz HP or 0.5Hz HP.
Your amplifier baseline (offset) is not adjusted properly. Please consult with application note 110 for details on how to make the required adjustments.
Smart Center Stand-alone, Portable Wireless Physiology Systems Smart Center Essentials entry-level system includes a Smart Center Device (BN-SMART with USB power cable), 2 or 3 Transmitters (BN-xxx-T with chargers); AcqKnowledge for Smart Center, and a case. Smart Center Enhanced adds a Logger (BN-LOGGER) and Basic Scripting License (ACK100W-BAS) to the Smart Center Device (BN-SMART with powercable), 3 […]
“Alan Macy: Scientist, Entrepreneur, and Visionary Community Builder” The Santa Barbara Independent honored Alan Macy as a Local Hero. “What makes Macy a Santa Barbara Independent Local Hero is the combination of vision, generosity, and persistence he has shown in bringing creative people together and in crafting innovative venues for them to collaborate. …For this, and for his […]
Good quality data is important for reliable physiological research. No longer restrained to wires or stationary devices, BIOPAC’s newly released Smart Center is a lightweight and compact system designed to acquire quality data. The portable, wireless device offers high fidelity physiological data, allowing for subjects to be untethered while being recorded. The system includes the […]