When the amplified signal for any given channel exceeds the range -10 to +10 volts, the signal will rail. You will see a straight line at -10 or + 10 volts (more likely the reading will be close 9.99 volts). The MP system is designed to work only in the range -10 to +10 volts. The signal could rail for several reasons (which are not exclusive):
The signal is too strong for the current gain setting on the amplifier. Try reducing the gain setting.
You are recording in DC mode when you should be recording in AC mode. If this is the case, make sure DC is off and a switch is set to 0.05Hz HP or 0.5Hz HP.
Your amplifier baseline (offset) is not adjusted properly. Please consult with application note 110 for details on how to make the required adjustments.
Early bird savings extended! T4: Tools, Trends, Techniques, and Technology Monday, July 24, 2017 – Wednesday, July 26, 2017 Join BIOPAC for three days of hands-on, small-group human physiology workshops, presented by renowned experts, on the beautiful University of California at Santa Barbara campus in Santa Barbara, California, USA. Who should attend Anyone who records and […]
BIOPAC provides software and hardware that allow for research teams to record and analyze respiration activity in physiological experimentation. Here are a few notable studies in monitoring respiration and cardiovascular activity. The coupling between peripheral microcirculation and slow breathing: The purpose of this study was to investigate the coupling of breathing movements and microcirculatory blood […]
CBS News | 60 Minutes | Brain Hacking APRIL 9, 2017| Why can’t we stop looking at our smartphones? And are the designers of the apps and content on them using brain science to keep us hooked? Anderson Cooper reports. BIOPAC users Larry D. Rosen, Ph.D. and Nancy A. Cheever, Ph.D. from California State […]