When the amplified signal for any given channel exceeds the range -10 to +10 volts, the signal will rail. You will see a straight line at -10 or + 10 volts (more likely the reading will be close 9.99 volts). The MP system is designed to work only in the range -10 to +10 volts. The signal could rail for several reasons (which are not exclusive):
The signal is too strong for the current gain setting on the amplifier. Try reducing the gain setting.
You are recording in DC mode when you should be recording in AC mode. If this is the case, make sure DC is off and a switch is set to 0.05Hz HP or 0.5Hz HP.
Your amplifier baseline (offset) is not adjusted properly. Please consult with application note 110 for details on how to make the required adjustments.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]