When the amplified signal for any given channel exceeds the range -10 to +10 volts, the signal will rail. You will see a straight line at -10 or + 10 volts (more likely the reading will be close 9.99 volts). The MP system is designed to work only in the range -10 to +10 volts. The signal could rail for several reasons (which are not exclusive):
The signal is too strong for the current gain setting on the amplifier. Try reducing the gain setting.
You are recording in DC mode when you should be recording in AC mode. If this is the case, make sure DC is off and a switch is set to 0.05Hz HP or 0.5Hz HP.
Your amplifier baseline (offset) is not adjusted properly. Please consult with application note 110 for details on how to make the required adjustments.
StimTracker Duo. StimTracker Quad MARK. SYNC. AUDIT. StimTracker Duo and Quad autonomously detect the onset of events to avoid operating system delays. The Quad adds direct TTL output, support for vocal response onset (voice key), more light sensors, and the option to install a Raspberry Pi board. Onset of Auditory Stimuli – Pass the audio via StimTracker, set […]
Recent developments regarding electric stimulation and mental processing have lead to interesting developments in psychophysiology and physiological sciences. The following studies look at developments in how we potentially care for and understand solutions to issues with mental health, as well as new ways to monitor electrical stimulus on participants. The ventromedial prefrontal cortex (vmPFC) is […]
“Alan Macy: Scientist, Entrepreneur, and Visionary Community Builder” The Santa Barbara Independent honored Alan Macy as a Local Hero. “What makes Macy a Santa Barbara Independent Local Hero is the combination of vision, generosity, and persistence he has shown in bringing creative people together and in crafting innovative venues for them to collaborate. …For this, and for his […]