When the amplified signal for any given channel exceeds the range -10 to +10 volts, the signal will rail. You will see a straight line at -10 or + 10 volts (more likely the reading will be close 9.99 volts). The MP system is designed to work only in the range -10 to +10 volts. The signal could rail for several reasons (which are not exclusive):
The signal is too strong for the current gain setting on the amplifier. Try reducing the gain setting.
You are recording in DC mode when you should be recording in AC mode. If this is the case, make sure DC is off and a switch is set to 0.05Hz HP or 0.5Hz HP.
Your amplifier baseline (offset) is not adjusted properly. Please consult with application note 110 for details on how to make the required adjustments.
The STMEPM Programmable Stimulation System for E-Prime allows a user to interface the STMISOLA Stimulator with E-Prime to control the stimulus frequency and stimulus intensity for real-time stimulus delivery changes based on a subject’s responses. It is also possible to hardcode the stimulus intensity levels in the presentation so that predefined stimulus levels are delivered […]
The new issue of Psychophysiology has a study “Quantifying rapid changes in cardiovascular state with a moving ensemble average” using MEAP for preprocessing/analysis and BIOPAC hardware and software for data collection, display, and storage. The proof of concept study demonstrates a viable method for the adoption of ICG measures across the field of psychophysiology. MEAP (moving ensemble […]
BIOPAC provides software and hardware that allow for research teams to record and analyze respiration activity in physiological experimentation. Here are a few notable studies covering MRI, ECG, and Laser Doppler Flow measurement. fMRI & Thermal Perception: The neural mechanisms underlying thermal perception (how hot or cold we perceive the temperature to be) have not been fully […]