When the amplified signal for any given channel exceeds the range -10 to +10 volts, the signal will rail. You will see a straight line at -10 or + 10 volts (more likely the reading will be close 9.99 volts). The MP system is designed to work only in the range -10 to +10 volts. The signal could rail for several reasons (which are not exclusive):
The signal is too strong for the current gain setting on the amplifier. Try reducing the gain setting.
You are recording in DC mode when you should be recording in AC mode. If this is the case, make sure DC is off and a switch is set to 0.05Hz HP or 0.5Hz HP.
Your amplifier baseline (offset) is not adjusted properly. Please consult with application note 110 for details on how to make the required adjustments.
Easily Schedule, Record, and Analyze Stellar Telemetry Data with AcqKnowledge Intuitive Animal Scheduler: Easy-to-configure calendar display—Select a subset or complete range of animals for long term recordings of conscious, unrestrained animals Customizable Display: View Data in Multiple Display modes simultaneously; Signal Conditioning Tools include filtering and artifact removal AcqKnowledge: Multi-Animal, Multi-Channel simultaneous automated data analysis. […]
The Human Anatomy & Physiology Society (HAPS) found that BIOPAC’s Student Lab System (BSL) and AcqKnowledge software were the number one ranked data acquisition and analysis systems for physiology experiments performed in introductory A&P labs. According to HAPS surveys conducted in 2013 and 2016, BIOPAC products are used and preferred by members in more labs than […]
Hardware and software packages from BIOPAC allow commercial consumer marketing teams to easily combine precise eye tracking data with physiological data recording. Here are a few notable studies recording physiology with eye tracking data: Predicting Advertising Success: While neurophysiological methods have become increasingly popular to better understand marketing phenomena among academics and practitioners, the success of […]