A notch filter is typically an aggressive filter that strongly attenuates the power over a narrow region of the spectrum. For example, a notch filter at 60 Hz / 50 Hz is used to filter out power line noise with minimal disruption to the rest of the signal.
All MP biopotential amplifier modules which contain a 50/60 Hz notch filter only engage the filter when the pass filter is also ON:
For amplifiers with a 35 Hz LPN low pass notch filter setting (ECG, EEG, EOG): the 50/60 Hz notch is only engaged when this filter is ON.
For amplifiers with a 100 Hz HPN high pass notch filter setting (EMG, ERS): the 50/60 HZ notch is only engaged when this filter is ON.
Plots of the frequency responses of these filters may be found in an Appendix in the MP Systems Hardware Guide (also available via the “Help” menu in AcqKnowledge).
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
MIT and Harvard researchers developed a new technique to expand ventilator capacity Individualized System for Augmenting Ventilator Efficacy (iSAVE): A Rapidly deployable system to expand ventilator capacity “The iSAVE is a rapidly deployable solution in response to the urgent shortage of mechanical ventilators for respiratory support during the COVID-19 pandemic. In contrast to previously described […]
New Ventilator Validation System Allows Device Manufacturers to Test Ventilator Pressure and Volume System supports rapid development and testing of hardware for COVID-19 treatment Goleta, California – March 24, 2020 BIOPAC Systems, Inc. announced a new Ventilator Validation System for engineering and device manufacturers to use when designing and testing new and existing ventilators for […]