AC mode indicates that a high-pass (HP) filter has been applied to the signal that you are recording. An HP filter removes the low-frequency components of a signal. The most common AC mode settings on BIOPAC amplifiers are 0.5 Hz HP and 0.05 Hz HP. In AC mode, the signal is centered around zero.
AC mode is useful in studying changes in the signal. AC mode is not appropriate for recording absolute measures or for detecting the baseline level of the signal.
The NIBP100E noninvasive blood pressure system provides continuous, beat‐to‐beat, blood pressure signal and values (Sys, Dia, MAP) and Pulse Rate (PR)*. The NIBP100E amplifier with CNAP™ technology is controlled by AcqKnowledge software (v5.0.7) for MP160 Research Systems. The system outputs a continuous blood pressure waveform that is similar to a direct arterial pressure waveform. The amplifier […]
BIOPAC provides software and hardware that allows research teams to study physiological measures related to stress. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and EEG/ERP. Moms Moderating Stress Can relaxation techniques help lower stress on both mother and fetus? Using an MP36R Data Acquisition system to record ECG and EDA, this study […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]