When using calibration gas tanks with a mixing chamber—such as the AFT15 or the GASSYS2 system—an interface is required between the gas tank barbed ports and the mixing chamber’s Luer lock female port. Here is a suggested method.
Acquire 3-6 feet of 1/8″ ID silicon tubing AND a 1/8” barb to male Luer connector (polycarbonate).
Assemble the male Luer-lock into one end of the silicon tubing.
Push the other open end of the tubing into the barb port of the gas regulator.
Interface the AFT20 filter disk end and without the Y-connector that comes with it, to the male Luer-lock connector.
Connect the AFT20 male Luer-lock to the female Luer-lock port of the mixing chamber.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Solving Physical Stress Issues for Surgeons Physical stress in the operating room is common for surgeons, considering how much time they spend in an upright standing position during surgical procedures. With procedures that require standing for several hours and average reported working times of 60-80 hours a week, surgeons seem to be at risk for musculoskeletal […]
BIOPAC combines eye tracking with mobile data logging and physiology data collection to run experiments in the lab or in the real world BIOPAC Systems, Inc. is pleased to announce new eye tracking integration for researchers who conduct experiments from mobile participants in diverse locations. ETVision (EYE-ETV) eye tracking glasses now integrate with BIOPAC wireless systems and software, […]