When using calibration gas tanks with a mixing chamber—such as the AFT15 or the GASSYS2 system—an interface is required between the gas tank barbed ports and the mixing chamber’s Luer lock female port. Here is a suggested method.
Acquire 3-6 feet of 1/8″ ID silicon tubing AND a 1/8” barb to male Luer connector (polycarbonate).
Assemble the male Luer-lock into one end of the silicon tubing.
Push the other open end of the tubing into the barb port of the gas regulator.
Interface the AFT20 filter disk end and without the Y-connector that comes with it, to the male Luer-lock connector.
Connect the AFT20 male Luer-lock to the female Luer-lock port of the mixing chamber.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]