When using calibration gas tanks with a mixing chamber—such as the AFT15 or the GASSYS2 system—an interface is required between the gas tank barbed ports and the mixing chamber’s Luer lock female port. Here is a suggested method.
Acquire 3-6 feet of 1/8″ ID silicon tubing AND a 1/8” barb to male Luer connector (polycarbonate).
Assemble the male Luer-lock into one end of the silicon tubing.
Push the other open end of the tubing into the barb port of the gas regulator.
Interface the AFT20 filter disk end and without the Y-connector that comes with it, to the male Luer-lock connector.
Connect the AFT20 male Luer-lock to the female Luer-lock port of the mixing chamber.
BSL System with MP36 and hardware for 11 guided lessons plus 14 dynamic lessons and student-designed Human Physiology experiments. Use the power of the BSL Human Physiology Teaching System to conduct the most widely studied responses in physiology teaching labs. Lessons target the circulatory system, respiratory system, muscular function, brain function, ANS, exercise physiology and […]
BIOPAC tools are used for studies in a variety of different fields and disciplines. AcqKnowledge software is one of our universally applicable tools, used by thousands of labs around the world and cited in thousands of peer reviewed publications, such as the following studies… The psychophysiology of guilt in healthy adults. Cognitive, Affective, & Behavioral Neuroscience, 1-18. […]
We had the opportunity to meet lots of researchers at our 4T|Phys Human Physiology Conference this month and demonstrate solutions for life science studies. We presented 39 unique skills sessions, including sessions on measuring blood pressure, recording HRV, and utilizing VR. These studies all apply these skills and further demonstrate how to apply our equipment… Real-Time Personalized Physiologically […]