When using calibration gas tanks with a mixing chamber—such as the AFT15 or the GASSYS2 system—an interface is required between the gas tank barbed ports and the mixing chamber’s Luer lock female port. Here is a suggested method.
Acquire 3-6 feet of 1/8″ ID silicon tubing AND a 1/8” barb to male Luer connector (polycarbonate).
Assemble the male Luer-lock into one end of the silicon tubing.
Push the other open end of the tubing into the barb port of the gas regulator.
Interface the AFT20 filter disk end and without the Y-connector that comes with it, to the male Luer-lock connector.
Connect the AFT20 male Luer-lock to the female Luer-lock port of the mixing chamber.
fNIRS is growing in popularity as a tool for studying human cognition including workload, memory, learning, attention and more. It is simpler than ever to incorporate into your experiments.
You will learn how to: – Operate fNIRS equipment – Record high quality data – Perform an experiment – Analyze data – Synchronize with other physiological signals – Synchronize with stimulus presentation
BIOPAC provides software and hardware that allows research teams to record and analyze numerous body signals for physiological experimentation. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and Respiration data. What Keeps Students More Engaged? The question has been brought to the forefront during COVID, what variations of online learning keep students more engaged? […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]