Precise real-time Oxygen Consumption (VO2) and Respiratory Exchange Ratio (RER) measurements can be performed with the MP150/100 System and the CO2100, O2100 and DA100 modules with some airflow accessories. When performing tests of this kind, many factors exist to confound the measurement. For example, it may be reasonable to think that the volume of expired air is always the same as inspired air, but this is only true when the volume of carbon dioxide expired is equal to the volume of oxygen consumed. Namely, the inspired and expired volumes are equal only when the RER equals one.
The Respiratory Exchange Ratio (RER) is defined as VCO2 produced divided by VO2 consumed (where V is volume). Accordingly, when performing these measurements, precise determination of inspired and expired volumes plus accurate gas concentration level recording is required. This application note focuses on the details of the setup and calculations to obtain precise VO2 and RER measurements from resting or exercising subjects.
The subjects breathe through a non-rebreathing “T” valve to inspire fresh air through a pneumotach (TSD107B) and expire to a mixing chamber for gas concentration (O2 and CO2) monitoring. When using a mixing chamber to average CO2 and O2 concentrations over several breaths, there is no performance degradation (due to the module gas concentration response time) when measuring these changes for arbitrarily high breathing rates. Accordingly, a mixing chamber is typically recommended for quick, accurate and easy metabolic analysis.
Additional accessories required, but not shown, include a calibration syringe (AFT6A) and calibrated gas mixture. The calibration syringe is used to calibrate the TSD107B and to flush out the mixing chamber for ambient gas concentration level calibration. The calibrated gas mixture is used to introduce known concentrations of CO2 and O2 to calibrate the upper and lower ranges of the gas measurement modules respectively.
fNIRS is growing in popularity as a tool for studying human cognition including workload, memory, learning, attention and more. It is simpler than ever to incorporate into your experiments.
You will learn how to: – Operate fNIRS equipment – Record high quality data – Perform an experiment – Analyze data – Synchronize with other physiological signals – Synchronize with stimulus presentation
BIOPAC provides software and hardware that allows research teams to record and analyze numerous body signals for physiological experimentation. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and Respiration data. What Keeps Students More Engaged? The question has been brought to the forefront during COVID, what variations of online learning keep students more engaged? […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]