Precise real-time Oxygen Consumption (VO2) and Respiratory Exchange Ratio (RER) measurements can be performed with the MP150/100 System and the CO2100, O2100 and DA100 modules with some airflow accessories. When performing tests of this kind, many factors exist to confound the measurement. For example, it may be reasonable to think that the volume of expired air is always the same as inspired air, but this is only true when the volume of carbon dioxide expired is equal to the volume of oxygen consumed. Namely, the inspired and expired volumes are equal only when the RER equals one.
The Respiratory Exchange Ratio (RER) is defined as VCO2 produced divided by VO2 consumed (where V is volume). Accordingly, when performing these measurements, precise determination of inspired and expired volumes plus accurate gas concentration level recording is required. This application note focuses on the details of the setup and calculations to obtain precise VO2 and RER measurements from resting or exercising subjects.
The subjects breathe through a non-rebreathing “T” valve to inspire fresh air through a pneumotach (TSD107B) and expire to a mixing chamber for gas concentration (O2 and CO2) monitoring. When using a mixing chamber to average CO2 and O2 concentrations over several breaths, there is no performance degradation (due to the module gas concentration response time) when measuring these changes for arbitrarily high breathing rates. Accordingly, a mixing chamber is typically recommended for quick, accurate and easy metabolic analysis.
Additional accessories required, but not shown, include a calibration syringe (AFT6A) and calibrated gas mixture. The calibration syringe is used to calibrate the TSD107B and to flush out the mixing chamber for ambient gas concentration level calibration. The calibrated gas mixture is used to introduce known concentrations of CO2 and O2 to calibrate the upper and lower ranges of the gas measurement modules respectively.
Basic principles of physiological data collection covers fundamentals of data collection and equipment configuration equipment to help you refine experiment protocols and avoid costly missteps. Frazer Findlay, CEO of BIOPAC, discusses common mistakes with physiology recording and shares his secrets for collecting great data. Topics include: How to prep a subject and where to place electrodes; Which type of electrodes work for different body signals; Analog to Digital Conversion; Sample rates (correct vs. incorrect); Filtering; and Scaling and calibration.Watch On Demand Now!
Recent psychophysiological studies feature BIOPAC’s MP Research series of Data Recorders and AcqKnowledge software. Creating Adaptive Assistants for helicopter crews based on Real-time Physiological Data In the Mercedes S-Class, biometric tools track a driver’s eyes, heart rate, and other physiological data, with one goal in mind: Making sure drivers are alert and safe on the […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]