When connecting the analog output sourcing from external devices to the MP100 or MP150, channel contention must be considered. To connect external device outputs to the MP100 or MP150:
Non-human subjects or only collecting data from external devices—If the MP System is only collecting signals from non-human subjects (via MP system amplifier modules) or if the MP System is only collecting data from external devices:
Connect external device output signal to an unused UIM100C input channel (1-16).
Human subjects—If the MP System is collecting signals from human subjects (via MP system amplifier modules), it’s important to isolate the external device output signal from the MP System input:
Connect external device output signal to an unused HLT100C input channel (1-16) via INISO.
Channel contention issues
If an analog channel is used on the UIM100C or HLT100C, make certain that two external devices do not use the same analog channel.
If amplifier modules are connected to the MP System then those amplifier modules must be set to a channel which is not used by external devices plugged into the UIM100C or HLT100C.
Two external device outputs and an ECG100C amplifier are connected to the MP150 system and connected to a human subject.
– Device one is a Noninvasive Blood Pressure (NIBP) monitor – Device two is an Electronic Scale
– The ECG100C module measures the subject’s electrocardiogram
In this case, to fully isolate the human subject:
Both the NIBP monitor and the Electronic scale outputs should be connected to the MP150 inputs via the HLT100C, using one INISO for each input channel.
The ECG100C should be snapped directly to the MP150 System and connected directly to the subject with the appropriate leads and electrodes.
Assuming the NIBP is connected via INISO to HLT100C channel 1 and the Electronic Scale is connected via INISO to HLT100C channel 2, then the ECG100C amplifier must be set to a channel between 3-16.
The ECG100C can’t use Channels 1 and 2 because other devices are using both of these channels.
MP160 data acquisition and analysis systems with AcqKnowledge software provide a flexible tool for life science research. MP160 data acquisition workstations include all hardware, software, cables, and guides you need to get started. MP160 Data Acquisition Hardware—Ethernet-ready data acquisition and analysis unit can record multiple channels with differing sample rates; record at speeds up to 400 kHz (aggregate). […]
BIOPAC provides software and hardware that allows engineers to study the body. Here are a few notable studies using BIOPAC equipment for Biomedical Engineering research. Rehabilitating the Rehab Industry Could physical therapy be done from the comforts of your own home? If possible, this would open the door for more people with all sorts of ailments to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]