When connecting the analog output sourcing from external devices to the MP100 or MP150, channel contention must be considered. To connect external device outputs to the MP100 or MP150:
Non-human subjects or only collecting data from external devices—If the MP System is only collecting signals from non-human subjects (via MP system amplifier modules) or if the MP System is only collecting data from external devices:
Connect external device output signal to an unused UIM100C input channel (1-16).
Human subjects—If the MP System is collecting signals from human subjects (via MP system amplifier modules), it’s important to isolate the external device output signal from the MP System input:
Connect external device output signal to an unused HLT100C input channel (1-16) via INISO.
Channel contention issues
If an analog channel is used on the UIM100C or HLT100C, make certain that two external devices do not use the same analog channel.
If amplifier modules are connected to the MP System then those amplifier modules must be set to a channel which is not used by external devices plugged into the UIM100C or HLT100C.
Two external device outputs and an ECG100C amplifier are connected to the MP150 system and connected to a human subject.
– Device one is a Noninvasive Blood Pressure (NIBP) monitor – Device two is an Electronic Scale
– The ECG100C module measures the subject’s electrocardiogram
In this case, to fully isolate the human subject:
Both the NIBP monitor and the Electronic scale outputs should be connected to the MP150 inputs via the HLT100C, using one INISO for each input channel.
The ECG100C should be snapped directly to the MP150 System and connected directly to the subject with the appropriate leads and electrodes.
Assuming the NIBP is connected via INISO to HLT100C channel 1 and the Electronic Scale is connected via INISO to HLT100C channel 2, then the ECG100C amplifier must be set to a channel between 3-16.
The ECG100C can’t use Channels 1 and 2 because other devices are using both of these channels.
BIOPAC is introducing Biopac Student Lab Home (BSL HOME)—a new product specifically geared towards teaching physiology labs remotely so students can still get hands-on, practical lab experience if social distancing and stay-at-home orders keep them out of the lab..
BSL HOME will allow your students to conduct physiology labs in the safety of their own home, dorm room, or any remote location.
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]
Comparing Muscle Movement and Involvement between Boxing Punches In the fast-paced, high-intensity combat sport of boxing, the right punch can win a fight. However, boxing is also a sport where opponents can become exhausted in mere minutes of action, and prioritizing their strength is key to winning a fight. This team recorded EMG on test […]