When connecting the analog output sourcing from external devices to the MP100 or MP150, channel contention must be considered. To connect external device outputs to the MP100 or MP150:
Non-human subjects or only collecting data from external devices—If the MP System is only collecting signals from non-human subjects (via MP system amplifier modules) or if the MP System is only collecting data from external devices:
Connect external device output signal to an unused UIM100C input channel (1-16).
Human subjects—If the MP System is collecting signals from human subjects (via MP system amplifier modules), it’s important to isolate the external device output signal from the MP System input:
Connect external device output signal to an unused HLT100C input channel (1-16) via INISO.
Channel contention issues
If an analog channel is used on the UIM100C or HLT100C, make certain that two external devices do not use the same analog channel.
If amplifier modules are connected to the MP System then those amplifier modules must be set to a channel which is not used by external devices plugged into the UIM100C or HLT100C.
Two external device outputs and an ECG100C amplifier are connected to the MP150 system and connected to a human subject.
– Device one is a Noninvasive Blood Pressure (NIBP) monitor
– Device two is an Electronic Scale
– The ECG100C module measures the subject’s electrocardiogram
In this case, to fully isolate the human subject:
Both the NIBP monitor and the Electronic scale outputs should be connected to the MP150 inputs via the HLT100C, using one INISO for each input channel.
The ECG100C should be snapped directly to the MP150 System and connected directly to the subject with the appropriate leads and electrodes.
Assuming the NIBP is connected via INISO to HLT100C channel 1 and the Electronic Scale is connected via INISO to HLT100C channel 2, then the ECG100C amplifier must be set to a channel between 3-16.
The ECG100C can’t use Channels 1 and 2 because other devices are using both of these channels.
Use this DSUB9 to RJ11 jack Output Adapter to map the analog output of an MP36 or MP36R to an RJ jack; allows stimulators designed with HLT-compatible connections to be connected to MP36/36R units. Control the STMTHERM, or other stimulators that have RJ11 input cables.
Recent Studies using BIOPACs MP-Series (MP160, MP150) data recorders, ECG Amplifiers, and BioNomadix Wireless recorders: Singing style to infants Mothers around the world sing to infants, presumably to regulate their mood and arousal. Lullabies and playsongs differ stylistically and have distinctive goals. Mothers sing lullabies to soothe and calm infants, and playsongs to engage and […]
Alex Dimov, BIOPAC Sales Executive, was interviewed at the Neuromarketing World Forum by Inside Marketing to discuss how virtual reality can improve marketing research. Alex explains how VR can be used to present virtual iterations of a product to save time and money, and notes that measuring physiological responses adds lots of information for product and messaging […]