Module SS31LA Impedance Cardiography Sensor with leads
Factors that Affect Bioimpedance Data
System bandwidth. Historical algorithms for ICG make assumptions about ICG bandwidth that are not commonly broadcast. For example, a 10 Hz LPF will act to limit the spikiness of the ICG waveform, so that it looks “clean” but if the algorithm implies peak dZ/dt as an input, then the estimated stroke volume will shift. However, this arrangement will continue to work well to provide relative changes in stroke volume.
Electrode type. The typical reference in old literature is circumferential tape; newer methods use paired spot electrodes.
Electrode placement. Location and position is hyper-critical. For example, TREV is far more stable than ICG, due to the simplicity of the internal forearm, compared to the complexity of the torso (with lungs, big vessels, etc.).
Participant position. Standing/supine: fluids shift in the torso, between these states, thus radically affecting the torso base impedance.
Bioimpedance methods to perform stroke volume and cardiac output measurements, via application of electrodes on the neck and torso, are considered by BIOPAC to be research and educational tools.
Historically, there have been numerous research efforts to measure stroke volumes and cardiac outputs using bioimpedance techniques. The performance of these systems are subject to evolving algorithms. New bioimpedance methods, such as TREV (TransRadial Electrical bioimpedance Velocimetry), are examples showing new promise in this area. Additionally, machine learning strategies are beginning to accommodate the variabilities of bioimpedance methods, due to electrode type, placement, body position, movement artifact and electrical signal filtering.
Research is ongoing, as bioimpedance techniques offer profound noninvasive advantages, compared to thermodilution and similar “gold-standard” historical methods for measuring stroke volume and cardiac output. BIOPAC is committed to continuing to offer educational and research solutions for the application of bioimpedance methods, to measure cardiovascular parameters, despite the present “state of the art” showing these measures to be generally more useful for determining relative changes versus absolute values.
This research system ships with an MP160 Data Acquisition with AMI100D Module Interface and AcqKnowledge software licensed for Workflow Automation. MP160 + AMI100D provides a flexible tool for life science researchers. The modular, powerful data acquisition system can be used with BIOPAC’s amplifiers and accessories AND with equipment you already have. To put together a system to […]
Whether using specialized equipment, performing repetitive tasks, or simply going about our daily work routine, the effects of design in our environment as well as the tools we use and the way we use them can impact our health, work performance, and quality of life. Here are a few studies that have recorded and analyzed […]