BIOPAC offers a noninvasive cardiac output amplifier (NICO100C) and an electrical bioimpedance amplifier (EBI100C).
NICO10C is more appropriate if you are primarily interested in recording cardiac output.
EBI100C is more suitable for general purpose bioimpedance measurements; however, EBI100C can also be used for cardiac output measurements.
The primary difference between the NICO100C and EBI100C is the signals that the amplifiers output. They both output two analog signals, but only one of the two signals is common to both amplifiers and that is the impedance signal.
Outputs: Z0 Magnitude of impedance and dZ/dt Derivative of impedance
The NICO100C module incorporates an internal, hardware-based, derivative function, which outputs dZ/dt simultaneously with Z (impedance magnitude). This internal derivative function also inverts the polarity of the dZ/dt signal so that it displays a positive-going peak, coincident with negative slopes indicated in Z, as per academic research convention.
Outputs: MAG Magnitude of impedance and PHS Phase of impedance
The EBI100C does not include an internal, hardware-based, derivative function for the Z (impedance magnitude) channel. If dZ/dt is required, it can be determined on an AcqKnowledge calculation channel. Channel scaling can be employed to specify the dZ/dt polarity desired.
High-performance mobile fNIR Imaging Systems are stand-alone functional brain imaging solutions for continuous NIR spectroscopy (NIRS). These top of the line systems provide cognitive function assessment and eliminate a great many of the drawbacks of a functional MRI. Mobile subjects are comfortable and can respond naturally in real-world situations to take a test or perform mobile […]
BIOPAC provides software and hardware that allows research teams to run autonomic function tests while collecting physiological data. Here are a few studies focusing on BIOPAC’s ability to record physiological signals during autonomic function testing. The Internet, Sleep, and Heart Rate Does internet usage lead to poorer sleep? Using a BIOPAC Data Acquisition System to measure cardiovascular […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]