BIOPAC offers a noninvasive cardiac output amplifier (NICO100C) and an electrical bioimpedance amplifier (EBI100C).
NICO10C is more appropriate if you are primarily interested in recording cardiac output.
EBI100C is more suitable for general purpose bioimpedance measurements; however, EBI100C can also be used for cardiac output measurements.
The primary difference between the NICO100C and EBI100C is the signals that the amplifiers output. They both output two analog signals, but only one of the two signals is common to both amplifiers and that is the impedance signal.
Outputs: Z0 Magnitude of impedance and dZ/dt Derivative of impedance
The NICO100C module incorporates an internal, hardware-based, derivative function, which outputs dZ/dt simultaneously with Z (impedance magnitude). This internal derivative function also inverts the polarity of the dZ/dt signal so that it displays a positive-going peak, coincident with negative slopes indicated in Z, as per academic research convention.
Outputs: MAG Magnitude of impedance and PHS Phase of impedance
The EBI100C does not include an internal, hardware-based, derivative function for the Z (impedance magnitude) channel. If dZ/dt is required, it can be determined on an AcqKnowledge calculation channel. Channel scaling can be employed to specify the dZ/dt polarity desired.
The NICO Smart Amplifier is specifically designed to measure thoracic impedance for performing noninvasive cardiac output measurements. The amplifier terminates in two connectors for impedance (Z) and its derivative (dZ), and has a 3 m cable that connects directly to an AMI100D Amplifier Input Module. Use with EL500 series disposable electrodes and LEAD131 for a four-spot […]
NIRS and CNAP—Pairing great tools for great data Cerebral hemoglobin concentration (blood mass in the brain) is frequently measured in the prefrontal cortex, and systemic arterial blood pressure is known to be a variable factor based on the hemoglobin concentration. A leg pressure cuff was used to constrict and release blood flow to create a […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]