BIOPAC offers a noninvasive cardiac output amplifier (NICO100C) and an electrical bioimpedance amplifier (EBI100C).
NICO10C is more appropriate if you are primarily interested in recording cardiac output.
EBI100C is more suitable for general purpose bioimpedance measurements; however, EBI100C can also be used for cardiac output measurements.
The primary difference between the NICO100C and EBI100C is the signals that the amplifiers output. They both output two analog signals, but only one of the two signals is common to both amplifiers and that is the impedance signal.
Outputs: Z0 Magnitude of impedance and dZ/dt Derivative of impedance
The NICO100C module incorporates an internal, hardware-based, derivative function, which outputs dZ/dt simultaneously with Z (impedance magnitude). This internal derivative function also inverts the polarity of the dZ/dt signal so that it displays a positive-going peak, coincident with negative slopes indicated in Z, as per academic research convention.
Outputs: MAG Magnitude of impedance and PHS Phase of impedance
The EBI100C does not include an internal, hardware-based, derivative function for the Z (impedance magnitude) channel. If dZ/dt is required, it can be determined on an AcqKnowledge calculation channel. Channel scaling can be employed to specify the dZ/dt polarity desired.
Register Today! T4: Tools, Trends, Techniques, and Technology Monday, July 24, 2017 – Wednesday, July 26, 2017 Join BIOPAC for three days of hands-on, small-group human physiology workshops, presented by renowned experts, on the beautiful University of California at Santa Barbara campus in Santa Barbara, California, USA. Who should attend Anyone who records and analyzes […]
The Mobita 32-Channel EEG recording system was used in research examining if physiological data can be used to predict truly random events that correspond to perceptual stimuli. Baumgart, et al. from the University of California Santa Barbara utilized a quantum random number generator (qRNG) to choose from three randomized conditions: light, sound, and no stimulus. […]
BIOPAC provides software and hardware that allow for research teams to record and analyze respiration activity in physiological experimentation. Here are a few notable studies in monitoring respiration and cardiovascular activity. The coupling between peripheral microcirculation and slow breathing: The purpose of this study was to investigate the coupling of breathing movements and microcirculatory blood […]