Heel/toe strike data is recorded on two analog input channels (one for heel strike, one for toe strike). Registered impulses on each channel simply reflect the timing marks associated with heel/toe strike contact during gait. The amplitude of each impulse is indicative of force measured at the time of strike. Although this amplitude value does not have an exact linear relationship to force, it is monotonically related. As force increases, amplitude increases. If precise force measurements are required, weights could be sequentially applied to each sensor to perform a rough calibration within a narrow operational range or an AcqKnowledge Expression channel could be used to linearize a heel/toe strike sensor over a wide operational range.
- Use two TSD111A transducers (four channels) to record from both feet simultaneously.
- Order additional sensors as RX111.
- See SS28LA for the MP36/36R equivalent with CH input connector (to dSub9)