|Implant Weight||2 week – 0.5 g|
2 month – 2.3 g
6 month – 4 g
|Footprint||2 week – 4 mm x 6 mm|
2 month – 7 mm x 9 mm
6 month – 7 mm x 12 mm
|Volume||2 week – 0.192 cubic centimeter|
2 month – 0.756 cubic centimeter
6 month – 1.344 cubic centimeter
|Input Impedance||5 MΩ|
|Bandwidth||0.1 100 Hz per channel|
|Implant Material||Medical Grade Epoxy|
This sensor ships with the default transmitter Gain setting; other Gain settings are available if requested before order is placed. AcqKnowledge software includes the scale settings for each transmitter gain option.
|EEG, ECoG, LFP – default||2000x||±1.0 mV range, 2.0 mVpp in = 4 V out|
|Status-Epilepticus||800x||±2.5 mV range, 5.0 mVpp in = 4 V out|
|LFP, EEG in rat or mouse pups||4000x||±0.5 mV range, 1.0 mVpp in = 4 V out|
Activate Sensors When You’re Ready To Use Them
Sensors are shipped deactivated and the required Sensor Activation & Test Utility (EPOCH-ACTI) lets you start the sensor battery when you’re ready to use the transmitter.
- The reusable activator is battery operated and can be used for any sensors that are ordered as deactivated.
- Transmitters must be activated within 6 months of shipment to ensure the full 2-month or 6-month active battery-life.
- The activator utility can also output two independent channels of data (sinewaves) for testing purposes.
(older part # from same line of EEG 2-channel common reference sensors: EPOCH-T1, EPOCH-T2, EPOCH-T6, EPOCH-P-EEG-SEN.)