|Implant Weight||2 week – 0.8 g|
2 month – 2.6 g
6 month – 4 g
|Footprint||2 week – 4 mm x 6 mm|
2 month – 7 mm x 9 mm
6 month – 7 mm x 12 mm
|Volume||2 week – 0.192 cubic centimeter|
2 month – 0.756 cubic centimeter
6 month – 1.344 cubic centimeter
|Input Impedance||5 MΩ|
|Bandwidth||0.1 100 Hz per channel|
|Implant Material||Medical Grade Epoxy|
This sensor ships with the default transmitter Gain setting; other Gain settings are available if requested before order is placed. AcqKnowledge software includes the scale settings for each transmitter gain option.
Activate Sensors When You’re Ready To Use Them
Sensors are shipped deactivated and the required Sensor Activation & Test Utility (EPOCH-ACTI) lets you start the sensor battery when you’re ready to use the transmitter.
- The reusable activator is battery operated and can be used for any sensors that are ordered as deactivated.
- Transmitters must be activated within 6 months of shipment to ensure the full 2-month or 6-month active battery-life.
- The activator utility can also output two independent channels of data (sinewaves) for testing purposes.
(older part # from same line of EEG 2-channel common reference sensors: EPOCH-T1, EPOCH-T2, EPOCH-T6, EPOCH-P-EEG-SEN.)