USB drivers for BIOPAC hardware are Microsoft certified and meet the USB communication standards at the time the hardware platform was released.
Under Windows XP and above, USB drivers are broadcast to all USB ports on the computer; under earlier operating systems, driver information is related only to the port that was used when software was installed.
We recommend that you use the latest drivers with your hardware. Click here to check for driver updates. To ensure correct driver installation for the system as specified by BIOPAC, you can check BIOPAC Support > Updates or the Microsoft site for all updates (and download or update from the site). BIOPAC driver update information is also broadcast via the Microsoft software update program. Please see the Driver Guide for detailed instructions.
Current Hardware Platforms
Four channel system: MP36 has been the current hardware platform since June 2008.
MP36 drivers (32-bit or 64-bit) are specifically for MP36 data acquisition units and are certified to USB 2.0 high speed.
Two channel system: MP45 has been the current hardware platform since December 2008.
MP45 drivers are specifically for MP45 data acquisition units. Most current USB adapter/controllers are compatible with the USB chip built into the MP45 unit; testing has confirmed that the following products are reliable: Intel 82801G (ICH7 Family) USB Universal Host Controller, Renesas Electronics USB 3.0 Host Controller (PCI), Renesas Electronics USB 3.0 Express Card Adapter, and StarTech USB 3.0 Express Card Adapter.
Older Hardware Platforms
MP35 drivers (32-bit or 64-bit) are specifically for MP35 data acquisition units and are certified to USB 1.1 and USB 2.0 full speed.
MP35 USB 3.0 Compatibility – The MP35 is a USB 2.0 full speed device and may not be compatible when directly connected to a USB 3.0 port. Certain USB 3.0 chipsets are known to be incompatible with the MP35, including but not limited to the Intel 7 Series/C210 chipset family used in newer Apple laptops and other laptop computers. With these incompatible chipsets you may encounter difficulty establishing communications with an MP35. It is recommended to use a USB 2.0 port (if available). If unavailable, try using a USB 2.0 hub connected to the USB 3.0 port and connecting the MP35 through the hub. Otherwise please contact BIOPAC Systems, Inc. about upgrading to MP36 or newer hardware which is certified compatible with USB 3.0.
USB1W is the USB adapter for MP100 and MP30 data acquisition units and is certified compatible only with USB 1.1.
USB1W USB 2.0 and 3.0 Compatibility – The USB1W may not be compatible with USB 2.0 and USB 3.0 controllers.
The ERS Evoked Response Smart Amplifier is an extremely low noise differential amplifier that accurately amplifies very small potentials, such as those associated with stimulus/response, startle response, ABR auditory brainstem response, somatosensory evoked response, signal-averaged recordings, and nerve conduction velocity recordings. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by […]
The mind and body work together in many ways and psychophysiology research provides insight to these connections. BIOPAC has a variety of systems to power your emotion, arousal, stress, cognitive assessment, group interaction, startle, and other psychophysiology research. Here are some recent studies that have used BIOPAC systems for psychophysiological research… Skin conductance, facial EMG, […]
For immediate release: January 4, 2023 Amiens, France and GOLETA, California, USA Seenel Imaging SAS and BIOPAC Systems, Inc. have formed a long-term global partnership to distribute MedelOpt®, Seenel Imaging’s groundbreaking wearable neuroimaging device for the neuroscience research market. Seenel Imaging, based in France, continues its international expansion by partnering with BIOPAC Systems, […]