USB drivers for BIOPAC hardware are Microsoft certified and meet the USB communication standards at the time the hardware platform was released.
Under Windows XP and above, USB drivers are broadcast to all USB ports on the computer; under earlier operating systems, driver information is related only to the port that was used when software was installed.
We recommend that you use the latest drivers with your hardware. Click here to check for driver updates. To ensure correct driver installation for the system as specified by BIOPAC, you can check BIOPAC Support > Updates or the Microsoft site for all updates (and download or update from the site). BIOPAC driver update information is also broadcast via the Microsoft software update program. Please see the Driver Guide for detailed instructions.
Current Hardware Platforms
Four channel system: MP36 has been the current hardware platform since June 2008.
MP36 drivers (32-bit or 64-bit) are specifically for MP36 data acquisition units and are certified to USB 2.0 high speed.
Two channel system: MP45 has been the current hardware platform since December 2008.
MP45 drivers are specifically for MP45 data acquisition units. Most current USB adapter/controllers are compatible with the USB chip built into the MP45 unit; testing has confirmed that the following products are reliable: Intel 82801G (ICH7 Family) USB Universal Host Controller, Renesas Electronics USB 3.0 Host Controller (PCI), Renesas Electronics USB 3.0 Express Card Adapter, and StarTech USB 3.0 Express Card Adapter.
Older Hardware Platforms
MP35 drivers (32-bit or 64-bit) are specifically for MP35 data acquisition units and are certified to USB 1.1 and USB 2.0 full speed.
MP35 USB 3.0 Compatibility – The MP35 is a USB 2.0 full speed device and may not be compatible when directly connected to a USB 3.0 port. Certain USB 3.0 chipsets are known to be incompatible with the MP35, including but not limited to the Intel 7 Series/C210 chipset family used in newer Apple laptops and other laptop computers. With these incompatible chipsets you may encounter difficulty establishing communications with an MP35. It is recommended to use a USB 2.0 port (if available). If unavailable, try using a USB 2.0 hub connected to the USB 3.0 port and connecting the MP35 through the hub. Otherwise please contact BIOPAC Systems, Inc. about upgrading to MP36 or newer hardware which is certified compatible with USB 3.0.
USB1W is the USB adapter for MP100 and MP30 data acquisition units and is certified compatible only with USB 1.1.
USB1W USB 2.0 and 3.0 Compatibility – The USB1W may not be compatible with USB 2.0 and USB 3.0 controllers.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
New Ventilator Validation System Allows Device Manufacturers to Test Ventilator Pressure and Volume System supports rapid development and testing of hardware for COVID-19 treatment Goleta, California – March 24, 2020 BIOPAC Systems, Inc. announced a new Ventilator Validation System for engineering and device manufacturers to use when designing and testing new and existing ventilators for […]
Space Flight Simulations Prevent Disasters Under psychological and physical stress, astronauts can put their mission and crew in danger. Looking into this issue, researchers created a potentially catastrophic spaceflight simulation using a virtual reality environment and recorded the heart rate of participants. Psychophysiological responses to the potentially disastrous flight simulation were recorded using a BIOPAC MP […]