USB drivers for BIOPAC hardware are Microsoft certified and meet the USB communication standards at the time the hardware platform was released.
Under Windows XP and above, USB drivers are broadcast to all USB ports on the computer; under earlier operating systems, driver information is related only to the port that was used when software was installed.
We recommend that you use the latest drivers with your hardware. Click here to check for driver updates. To ensure correct driver installation for the system as specified by BIOPAC, you can check BIOPAC Support > Updates or the Microsoft site for all updates (and download or update from the site). BIOPAC driver update information is also broadcast via the Microsoft software update program. Please see the Driver Guide for detailed instructions.
Current Hardware Platforms
Four channel system: MP36 has been the current hardware platform since June 2008.
MP36 drivers (32-bit or 64-bit) are specifically for MP36 data acquisition units and are certified to USB 2.0 high speed.
Two channel system: MP45 has been the current hardware platform since December 2008.
MP45 drivers are specifically for MP45 data acquisition units. Most current USB adapter/controllers are compatible with the USB chip built into the MP45 unit; testing has confirmed that the following products are reliable: Intel 82801G (ICH7 Family) USB Universal Host Controller, Renesas Electronics USB 3.0 Host Controller (PCI), Renesas Electronics USB 3.0 Express Card Adapter, and StarTech USB 3.0 Express Card Adapter.
Older Hardware Platforms
MP35 drivers (32-bit or 64-bit) are specifically for MP35 data acquisition units and are certified to USB 1.1 and USB 2.0 full speed.
MP35 USB 3.0 Compatibility – The MP35 is a USB 2.0 full speed device and may not be compatible when directly connected to a USB 3.0 port. Certain USB 3.0 chipsets are known to be incompatible with the MP35, including but not limited to the Intel 7 Series/C210 chipset family used in newer Apple laptops and other laptop computers. With these incompatible chipsets you may encounter difficulty establishing communications with an MP35. It is recommended to use a USB 2.0 port (if available). If unavailable, try using a USB 2.0 hub connected to the USB 3.0 port and connecting the MP35 through the hub. Otherwise please contact BIOPAC Systems, Inc. about upgrading to MP36 or newer hardware which is certified compatible with USB 3.0.
USB1W is the USB adapter for MP100 and MP30 data acquisition units and is certified compatible only with USB 1.1.
USB1W USB 2.0 and 3.0 Compatibility – The USB1W may not be compatible with USB 2.0 and USB 3.0 controllers.
Easily Schedule, Record, and Analyze Stellar Telemetry Data with AcqKnowledge Intuitive Animal Scheduler: Easy-to-configure calendar display—Select a subset or complete range of animals for long term recordings of conscious, unrestrained animals Customizable Display: View Data in Multiple Display modes simultaneously; Signal Conditioning Tools include filtering and artifact removal AcqKnowledge: Multi-Animal, Multi-Channel simultaneous automated data analysis. […]
The Human Anatomy & Physiology Society (HAPS) found that BIOPAC’s Student Lab System (BSL) and AcqKnowledge software were the number one ranked data acquisition and analysis systems for physiology experiments performed in introductory A&P labs. According to HAPS surveys conducted in 2013 and 2016, BIOPAC products are used and preferred by members in more labs than […]
Hardware and software packages from BIOPAC allow commercial consumer marketing teams to easily combine precise eye tracking data with physiological data recording. Here are a few notable studies recording physiology with eye tracking data: Predicting Advertising Success: While neurophysiological methods have become increasingly popular to better understand marketing phenomena among academics and practitioners, the success of […]