If the signal from the amplifier is not on zero, it is possible to remove the offset by using the Zero Adj screw at the top of the amplifier.
Make sure that the amplifier is set to the correct mode (AC/DC), the filters are adjusted correctly for the signal, and the electrodes or transducer is connected.
Slowly turn the Zero Adj screw to adjust the zero position.
Optional: Adjust the vertical scale to see the small changes that occur when you turn the screw.
Note that if the gain is set too high the signal could rail at +10 or -10 Volts. It is sometimes easier to perform the adjustment if the gain is reduced. Turning the screw clockwise will move the signal up.
The zero position should not require constant adjustment. The amplifiers are very stable, and once the baseline has been set, it should not require further adjustment.
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
Studying Cognitive Activity with fNIR! The age-old game of chess is known as a thinking man’s game, where foresight, critical thinking, and strategy are required to become a good chess player, and the terms “genius” or “brilliance” are bestowed upon the chess masters of the world. As such, chess is an excellent cognitive tool to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]