If the signal from the amplifier is not on zero, it is possible to remove the offset by using the Zero Adj screw at the top of the amplifier.
Make sure that the amplifier is set to the correct mode (AC/DC), the filters are adjusted correctly for the signal, and the electrodes or transducer is connected.
Slowly turn the Zero Adj screw to adjust the zero position.
Optional: Adjust the vertical scale to see the small changes that occur when you turn the screw.
Note that if the gain is set too high the signal could rail at +10 or -10 Volts. It is sometimes easier to perform the adjustment if the gain is reduced. Turning the screw clockwise will move the signal up.
The zero position should not require constant adjustment. The amplifiers are very stable, and once the baseline has been set, it should not require further adjustment.
Smart shirt simultaneously acquires ECG and Respiration data while subjects roam freely The lightweight, comfortable BioNomadix BioShirt contains a respiration sensor and fabric electrodes to wirelessly record both respiration and ECG while ambulatory subjects move freely and perform tasks in short or long-term studies, in the lab, or in the real world. The shirt connects to a wireless BioNomadix BN-RSPEC Respiration […]
Research posters presented at BIOPAC’s T4 physiology conference (Tools, Trends, Techniques, and Technology) exhibited leading physiological research utilizing BIOPAC’s technology. Research posters presented unique advances in neuromarketing and health research. The T4 conference, held at UCSB Monday, July 24, 2017 through Wednesday, July 26, 2017, showcased the latest trends in the Tech industry’s physiological research. […]
ABC News | Good Morning America | Excessive Cell Phone Use JULY 31, 2017 | Excessive cell phone use may cause anxiety, experts warn BIOPAC user Nancy A. Cheever, Ph.D. from California State University, Dominguez Hills was featured on Good Morning America. Her study investigated whether spending too much time on your phone may be causing people to feel […]