If the signal from the amplifier is not on zero, it is possible to remove the offset by using the Zero Adj screw at the top of the amplifier.
Make sure that the amplifier is set to the correct mode (AC/DC), the filters are adjusted correctly for the signal, and the electrodes or transducer is connected.
Slowly turn the Zero Adj screw to adjust the zero position.
Optional: Adjust the vertical scale to see the small changes that occur when you turn the screw.
Note that if the gain is set too high the signal could rail at +10 or -10 Volts. It is sometimes easier to perform the adjustment if the gain is reduced. Turning the screw clockwise will move the signal up.
The zero position should not require constant adjustment. The amplifiers are very stable, and once the baseline has been set, it should not require further adjustment.
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
BIOPAC provides software and hardware that allows research teams to run autonomic function tests while collecting physiological data. Here are a few studies focusing on BIOPAC’s ability to record physiological signals during autonomic function testing. The Internet, Sleep, and Heart Rate Does internet usage lead to poorer sleep? Using a BIOPAC Data Acquisition System to measure cardiovascular […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]