If the signal from the amplifier is not on zero, it is possible to remove the offset by using the Zero Adj screw at the top of the amplifier.
Make sure that the amplifier is set to the correct mode (AC/DC), the filters are adjusted correctly for the signal, and the electrodes or transducer is connected.
Slowly turn the Zero Adj screw to adjust the zero position.
Optional: Adjust the vertical scale to see the small changes that occur when you turn the screw.
Note that if the gain is set too high the signal could rail at +10 or -10 Volts. It is sometimes easier to perform the adjustment if the gain is reduced. Turning the screw clockwise will move the signal up.
The zero position should not require constant adjustment. The amplifiers are very stable, and once the baseline has been set, it should not require further adjustment.
New series of unshielded 45 cm electrode leads with thin cable; suitable for facial EMG and other areas where lighter, shorter lead cables are required. Use for female Touchproof connectors to pinch clip connectors—connect electrodes to either a C-series amplifier or an MEC. The pinch connectors are light weight and the lead cable tinsel wire […]
The new issue of Psychophysiology has a study “Quantifying rapid changes in cardiovascular state with a moving ensemble average” using MEAP for preprocessing/analysis and BIOPAC hardware and software for data collection, display, and storage. The proof of concept study demonstrates a viable method for the adoption of ICG measures across the field of psychophysiology. MEAP (moving ensemble […]
BIOPAC provides software and hardware that allow for research teams to record and analyze respiration activity in physiological experimentation. Here are a few notable studies covering MRI, ECG, and Laser Doppler Flow measurement. fMRI & Thermal Perception: The neural mechanisms underlying thermal perception (how hot or cold we perceive the temperature to be) have not been fully […]