If the signal from the amplifier is not on zero, it is possible to remove the offset by using the Zero Adj screw at the top of the amplifier.
Make sure that the amplifier is set to the correct mode (AC/DC), the filters are adjusted correctly for the signal, and the electrodes or transducer is connected.
Slowly turn the Zero Adj screw to adjust the zero position.
Optional: Adjust the vertical scale to see the small changes that occur when you turn the screw.
Note that if the gain is set too high the signal could rail at +10 or -10 Volts. It is sometimes easier to perform the adjustment if the gain is reduced. Turning the screw clockwise will move the signal up.
The zero position should not require constant adjustment. The amplifiers are very stable, and once the baseline has been set, it should not require further adjustment.
Learn how to record event-related or endogenous evoked potentials (ERP).
You will learn about: – Introduction to ERPs – Theory and guidelines for recording ERPs – How to record great ERPs data – Wired and wireless wearable EEG-fNIRS bimodal response to oddball paradigm – How to mismatch negativity in a virtual world – How to prepare data for analysis – Workflow Automation Tools
The mind and body work together in many ways and psychophysiology research provides insight to these connections. BIOPAC has a variety of systems to power your emotion, arousal, stress, cognitive assessment, group interaction, startle, and other psychophysiology research. Here are some recent studies that have used BIOPAC systems for psychophysiological research… Skin conductance, facial EMG, […]
For immediate release: January 4, 2023 Amiens, France and GOLETA, California, USA Seenel Imaging SAS and BIOPAC Systems, Inc. have formed a long-term global partnership to distribute MedelOpt®, Seenel Imaging’s groundbreaking wearable neuroimaging device for the neuroscience research market. Seenel Imaging, based in France, continues its international expansion by partnering with BIOPAC Systems, […]