If the signal from the amplifier is not on zero, it is possible to remove the offset by using the Zero Adj screw at the top of the amplifier.
Make sure that the amplifier is set to the correct mode (AC/DC), the filters are adjusted correctly for the signal, and the electrodes or transducer is connected.
Slowly turn the Zero Adj screw to adjust the zero position.
Optional: Adjust the vertical scale to see the small changes that occur when you turn the screw.
Note that if the gain is set too high the signal could rail at +10 or -10 Volts. It is sometimes easier to perform the adjustment if the gain is reduced. Turning the screw clockwise will move the signal up.
The zero position should not require constant adjustment. The amplifiers are very stable, and once the baseline has been set, it should not require further adjustment.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Recent physiological studies feature BIOPAC’s Wireless BioNomadix Data Recorders Using Metronomes and Rhythmic stimulation to help people with Parkinson’s disease walk For those living with Parkinson’s disease, even simple tasks can take too long to accomplish properly, such as crossing a room or picking up an object. In this study, participants with diagnosed cases of […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]