n/a: strip electrode; Ag laminated on med. grade cloth
n/a: dry electrode—use your own choice of gel
For electrical stimulation, generally, the lower the electrode salt content, the less skin irritation the subject will experience; this is because fewer salt ions are driven into skin during stimulation. This strategy puts more demand on the stimulator, because skin impedances are higher with low salt electrodes. There is no one size fits all requirement. For example, it’s possible to help mitigate the undesirable effects of higher salt content electrodes by stimulating the subject with bipolar pulses, instead of unipolar pulses.
The salt content in all SOLID gel electrodes from BIOPAC is 4%. This universal gel can be used short and long term, and is suitable for adult and infants.
SOLID gels are gentler on the skin than WET gels at the same salt concentration. This is due to the higher water content in the WET gels that make them react faster and more aggressively on the skin.
If there are two (2) electrodes with 4% salt , one (1) WET gel and one (1) SOLID gel, the WET gel would be more reactive/aggressive.
The salt content in WET gel electrodes from BIOPAC varies:
10% is used for short term applications such as Resting ECG or Stress Test.
7% is a more universal gel and can be used for short and—for most subjects, though some react—long term
3% is a monitoring gel used for more than 24 hours.
1% is a sensitive skin gel, common (infants/elders).
0.5% in EDA/GSR electrodes.
We do not recommend for applications running more than 24 hours.
Possible skin irritation can result from the gel or the adhesive on the body of the electrode.
EL503 gel and adhesive are aggressive
EL502 uses a non-aggressive gel (solid gels are always less aggressive than wet gels with similar salt concentrations) with an aggressive adhesive.
Note—About 2 % of the population will react to any adhesives and gels put on a skin, regardless of composition or concentration.
Electrodermal Activity (EDA) is an objective index for understanding emotional states. Researchers measure EDA to gain insights on psychological or physiological response, leading them down the path of scientific discovery. Because EDA is so important to research, BIOPAC has created tools to optimize event-related EDA analysis. Topics include: how to identify skin conductance responses, convert stim events from E-Prime SuperLab, Presentation, Vizard, etc., into AcqKnowledge events that can used for analysis, and score skin conductance responses using the AcqKnowledge automated EDA analysis tool.
NIRS and CNAP—Pairing great tools for great data Cerebral hemoglobin concentration (blood mass in the brain) is frequently measured in the prefrontal cortex, and systemic arterial blood pressure is known to be a variable factor based on the hemoglobin concentration. A leg pressure cuff was used to constrict and release blood flow to create a […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]