n/a: strip electrode; Ag laminated on med. grade cloth
n/a: dry electrode—use your own choice of gel
For electrical stimulation, generally, the lower the electrode salt content, the less skin irritation the subject will experience; this is because fewer salt ions are driven into skin during stimulation. This strategy puts more demand on the stimulator, because skin impedances are higher with low salt electrodes. There is no one size fits all requirement. For example, it’s possible to help mitigate the undesirable effects of higher salt content electrodes by stimulating the subject with bipolar pulses, instead of unipolar pulses.
The salt content in all SOLID gel electrodes from BIOPAC is 4%. This universal gel can be used short and long term, and is suitable for adult and infants.
SOLID gels are gentler on the skin than WET gels at the same salt concentration. This is due to the higher water content in the WET gels that make them react faster and more aggressively on the skin.
If there are two (2) electrodes with 4% salt , one (1) WET gel and one (1) SOLID gel, the WET gel would be more reactive/aggressive.
The salt content in WET gel electrodes from BIOPAC varies:
10% is used for short term applications such as Resting ECG or Stress Test.
7% is a more universal gel and can be used for short and—for most subjects, though some react—long term
3% is a monitoring gel used for more than 24 hours.
1% is a sensitive skin gel, common (infants/elders).
0.5% in EDA/GSR electrodes.
We do not recommend for applications running more than 24 hours.
Possible skin irritation can result from the gel or the adhesive on the body of the electrode.
EL503 gel and adhesive are aggressive
EL502 uses a non-aggressive gel (solid gels are always less aggressive than wet gels with similar salt concentrations) with an aggressive adhesive.
Note—About 2 % of the population will react to any adhesives and gels put on a skin, regardless of composition or concentration.
fNIR Imaging Systems are stand-alone functional brain imaging solutions for continuous NIR spectroscopy (NIRS). They provide an in-lab cognitive function assessment and eliminate a great many of the drawbacks of a functional MRI. fNIR optical imaging technology measures neural activity and hemodynamic response in the prefrontal cortex. Subjects wear an fNIR sensor (IR light sources […]
Alan Macy Answers Proust Questionnaire BIOPAC co-founder and R&D Director, Alan Macy, was interviewed recently by Roger Durling for the Santa Barbara Independent. The article, “The S.B. Questionnaire: Alan Macy,” provided a background excerpt of the Santa Barbara entrepreneurs’ past as well as his answers to the Proust Questionnaire. The biomedical technology developed at BIOPAC reflects an ambitious […]
Physiological researchers have been studying the lower body through examining impacts of training, movement instruction, and innovative research measurements on an individual’s well being. The following articles represent recent developments in physiological research, moving our understanding of lower limbs and the body as a whole forward, one step at a time. The Importance of Form Improper […]