The following procedure will allow you to paste Append markers with the data. Use the Analysis->Find Cycle function to place markers in the data at the time when the append markers occurred.
Go to Analysis->Find Cycle.
Choose Locate Cycles from events.
choose General->Append. Uncheck Match Pairs of events only.
Click on Selection tab.
Make sure it reads Starting event plus 0 seconds for left edge and Ending event + 0 for the Right Edge.
Click on Output Tab.
Under Events, check the output events box. Event 1 should be set as follows: Left Edge, At Edge, Waveform onset (or whatever other marker you want to use to place at the append marker locations); output channel should be set to one of the data channels. Set Event 2 to None.
Click Find All Cycles. New markers will be placed at the locations corresponding to the append markers and can be copied along with the data.
If you only have a few Append markers that you want to paste, the following option might prove a little easier.
Open up the event palette from Display->Show->Event Palette.
Select each Append marker.
From Selected Event Channel, change from Global to one of the analog channels. The append markers will now appear on that analog channel and can be copied and pasted.
The STMEPM Programmable Stimulation System for E-Prime allows a user to interface the STMISOLA Stimulator with E-Prime to control the stimulus frequency and stimulus intensity for real-time stimulus delivery changes based on a subject’s responses. It is also possible to hardcode the stimulus intensity levels in the presentation so that predefined stimulus levels are delivered […]
Academics and professionals in the field of human physiology gathered at UCSB for the inaugural T4 Physiology Conference. BIOPAC coordinated the three day conference held in late July, which attracted attendees from 26 different countries. The attendees gathered to learn the latest tools, trends, techniques and technology used in life science measurement and interpretation. The […]
Industry reporting firm medGadget recognized BIOPAC as a major market competitor in the global Blood Flow Measurement Devices market. BIOPAC hardware solutions for blood flow measurement include laser Doppler flowmetry and extensive online and offline analysis of raw blood flow data to take an array of calculations and measurements, such as Pulsatile max, Pulsatile Min, […]