Here are some suggestions for overcoming the problem of stimulation artifact when applying a stimulation voltage to a muscle.
Have the ground between the stimulating electrodes and the recording electrodes.
The recording electrodes should be placed orthogonal to the stimulating electrodes.
Apply the highest possible low pass filter on the amplifier. This will allow the amplifier to recover faster.
During analysis, the find peak function should be used to look for EMG activity between the electrical stimuli. For example, if the stimuli are 100 msec apart and it takes 10 msec for the amplifier to recover from the stimulus a find peak from 10 to 99 msec after the interval can be used.
The stimulus artifact may never be removed completely because it is so intense. But this will not be much of an issue since the EMG signal you need for analysis falls between the spikes in the signal caused by the stimulator.
High-performance mobile fNIR Imaging Systems are stand-alone functional brain imaging solutions for continuous NIR spectroscopy (NIRS). These top of the line systems provide cognitive function assessment and eliminate a great many of the drawbacks of a functional MRI. Mobile subjects are comfortable and can respond naturally in real-world situations to take a test or perform mobile […]
Alex Dimov, BIOPAC Sales Executive, was interviewed at the Neuromarketing World Forum by Inside Marketing to discuss how virtual reality can improve marketing research. Alex explains how VR can be used to present virtual iterations of a product to save time and money, and notes that measuring physiological responses adds lots of information for product and messaging […]
February 14, 2019 BIOPAC Systems, Inc. Goleta, CA USA New “Smart Amplifiers” make it easier than ever for life science researchers to get great data Fast Setup & High Performance Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level voltage connection to the data acquisition system and reduces noise artifact. […]