Here are some suggestions for overcoming the problem of stimulation artifact when applying a stimulation voltage to a muscle.
Have the ground between the stimulating electrodes and the recording electrodes.
The recording electrodes should be placed orthogonal to the stimulating electrodes.
Apply the highest possible low pass filter on the amplifier. This will allow the amplifier to recover faster.
During analysis, the find peak function should be used to look for EMG activity between the electrical stimuli. For example, if the stimuli are 100 msec apart and it takes 10 msec for the amplifier to recover from the stimulus a find peak from 10 to 99 msec after the interval can be used.
The stimulus artifact may never be removed completely because it is so intense. But this will not be much of an issue since the EMG signal you need for analysis falls between the spikes in the signal caused by the stimulator.
Learn how to collect and interpret facial expressions data with facial coding software called FaceReader, and how to synchronize with other biometric response. FaceReader is an emotion reading software that analyzes facial movements to classify subject response. Facial expressions can be classified as happy, sad, scared, disgusted, surprised, angry, contempt, and neutral.
BIOPAC provides software and hardware that allow for research teams to record and analyze respiration activity in physiological experimentation. Here are a few notable studies covering HRV, EMG, and EEG measurement. DON’T STRESS (Jing Wei, Hong Luo, Si J. Wu, Paul P. Zheng, Genyue Fu, and Kang Lee) Humans encounter various stressful situations everyday at work, home, and school. Such stress when experienced at high degrees and/or […]
Alan Macy Answers Proust Questionnaire BIOPAC co-founder and R&D Director, Alan Macy, was interviewed recently by Roger Durling for the Santa Barbara Independent. The article, “The S.B. Questionnaire: Alan Macy,” provided a background excerpt of the Santa Barbara entrepreneurs’ past as well as his answers to the Proust Questionnaire. The biomedical technology developed at BIOPAC reflects an ambitious […]