Here are some suggestions for overcoming the problem of stimulation artifact when applying a stimulation voltage to a muscle.
Have the ground between the stimulating electrodes and the recording electrodes.
The recording electrodes should be placed orthogonal to the stimulating electrodes.
Apply the highest possible low pass filter on the amplifier. This will allow the amplifier to recover faster.
During analysis, the find peak function should be used to look for EMG activity between the electrical stimuli. For example, if the stimuli are 100 msec apart and it takes 10 msec for the amplifier to recover from the stimulus a find peak from 10 to 99 msec after the interval can be used.
The stimulus artifact may never be removed completely because it is so intense. But this will not be much of an issue since the EMG signal you need for analysis falls between the spikes in the signal caused by the stimulator.
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
BIOPAC provides software and hardware that allows research teams to record and analyze numerous body signals for physiological experimentation. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and Respiration data. What Keeps Students More Engaged? The question has been brought to the forefront during COVID, what variations of online learning keep students more engaged? […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]