BIOPAC offers a variety of setups for recording electrodermal activity, including
Reusable: skin conductance transducer (MP150+TSD203 or MP36+SS3LA) and electrodermal gel (GEL101).
Disposable: MP150+EDA100C amp with 2 x LEAD110A or MP36+SS57LA EDA lead let plus pre-gelled EDA electrodes (EL507).
Wireless or Logged: BN-PPGED amplifier or BN-LOGGER+BN-PPGED-T transmitter plus BN-EDA-LEAD2 and pre-gelled EDA electrodes EL507.
MRI: EDA100C-MRI amplifier with MR-conditional leads (2 x LEAD108B 15 cm or LEAD108C 30 cm), radio-translucent electrodes (2 x EL509 dry electrodes), and electrodermal gel (GEL101).
The following general guidelines will improve the quality of data from any setup:
Never abrade the skin.
Note: For MRI setups, lightly abrading the skin may sometimes help if the subject has particularly calloused fingers. Still, this is not recommended and should be used as a last resort only. See MRI Guidelines for additional information.
If you clean the electrode site, do it with water for all subjects.
Make sure you have enough gel on the electrodes. (Dry disposable electrodes will result in a deteriorated signal. Follow the storage guidelines for the electrodes to prevent them from drying out)
Use isotonic gel. (If using disposable electrodes and they appear dry you can still use them by applying isotonic gel as long as the dry gel on the electrodes has not crystallized)
Attach the electrodes 5-10 minutes before recording.
To confirm whether they are making good contact ask the subject to take a deep breath and hold it for a few seconds. This will usually result in a skin conductance response (see Deep breath and SCR/EDA (GSR) increase).
SMI Eye Tracking Glasses 2W with HD Scene Camera Now researchers can easily record, synchronize, and analyze biometric data and mobile eye tracking data during interactive tasks using SMI Eye Tracking Glasses 2 Wireless (ETG 2w) by SensoMotoric Instruments (SMI) BIOPAC wireless, wearable BioNomadix Logger, and BIOPAC AcqKnowledge software. The ETG 2w eye tracking glasses […]
Hardware and software packages from BIOPAC allow commercial consumer marketing teams to easily combine precise eye tracking data with physiological data recording. Here are a few notable studies recording physiology with eye tracking data: Predicting Advertising Success: While neurophysiological methods have become increasingly popular to better understand marketing phenomena among academics and practitioners, the success of […]
BIOPAC ECG solutions have been used in several research studies, including many that pertain to consumer behavior or neuromarketing. We have collected a few notable articles recording ECG in consumer research: Impact of Malevolent Business Practices: With information on potentially suspect business practices more widely available than ever, it is important to understand what this […]