BIOPAC offers a variety of setups for recording electrodermal activity, including
Reusable: skin conductance transducer (MP150+TSD203 or MP36+SS3LA) and electrodermal gel (GEL101).
Disposable: MP150+EDA100C amp with 2 x LEAD110A or MP36+SS57LA EDA lead let plus pre-gelled EDA electrodes (EL507).
Wireless or Logged: BN-PPGED amplifier or BN-LOGGER+BN-PPGED-T transmitter plus BN-EDA-LEAD2 and pre-gelled EDA electrodes EL507.
MRI: EDA100C-MRI amplifier with MR-conditional leads (2 x LEAD108B 15 cm or LEAD108C 30 cm), radio-translucent electrodes (2 x EL509 dry electrodes), and electrodermal gel (GEL101).
The following general guidelines will improve the quality of data from any setup:
Never abrade the skin.
Note: For MRI setups, lightly abrading the skin may sometimes help if the subject has particularly calloused fingers. Still, this is not recommended and should be used as a last resort only. See MRI Guidelines for additional information.
If you clean the electrode site, do it with water for all subjects.
Make sure you have enough gel on the electrodes. (Dry disposable electrodes will result in a deteriorated signal. Follow the storage guidelines for the electrodes to prevent them from drying out)
Use isotonic gel. (If using disposable electrodes and they appear dry you can still use them by applying isotonic gel as long as the dry gel on the electrodes has not crystallized)
Attach the electrodes 5-10 minutes before recording.
To confirm whether they are making good contact ask the subject to take a deep breath and hold it for a few seconds. This will usually result in a skin conductance response (see Deep breath and SCR/EDA (GSR) increase).
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]