The best solution is to avoid the ECG artifact in the first place. The problem is typically solved by adjusting the placement of the electrodes. Place the EMG electrodes so that they are NOT aligned with the axis of electrical activity of the heart. Placing both EMG electrodes (Vin+ and Vin- ) on the same side of the body will help.
ECG artifact can be removed from EMG data that has been recorded by applying a 50Hz High pass FIR filter. The number of coefficients should be greater than or equal to two times the sampling rate, divided by the lowest cutoff frequency specified. For example, if running a low pass filter at 1Hz on data sampled at 100Hz, choose at least (2 X 100/1) or 200 coefficients in the filter. Additional coefficients will improve the response. Use a number four times higher.
The NIBP100E noninvasive blood pressure system provides continuous, beat‐to‐beat, blood pressure signal and values (Sys, Dia, MAP) and Pulse Rate (PR)*. The NIBP100E amplifier with CNAP™ technology is controlled by AcqKnowledge software (v5.0.7) for MP160 Research Systems. The system outputs a continuous blood pressure waveform that is similar to a direct arterial pressure waveform. The amplifier […]
BIOPAC provides software and hardware that allows research teams to study physiological measures related to stress. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and EEG/ERP. Moms Moderating Stress Can relaxation techniques help lower stress on both mother and fetus? Using an MP36R Data Acquisition system to record ECG and EDA, this study […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]