The best solution is to avoid the ECG artifact in the first place. The problem is typically solved by adjusting the placement of the electrodes. Place the EMG electrodes so that they are NOT aligned with the axis of electrical activity of the heart. Placing both EMG electrodes (Vin+ and Vin- ) on the same side of the body will help.
ECG artifact can be removed from EMG data that has been recorded by applying a 50Hz High pass FIR filter. The number of coefficients should be greater than or equal to two times the sampling rate, divided by the lowest cutoff frequency specified. For example, if running a low pass filter at 1Hz on data sampled at 100Hz, choose at least (2 X 100/1) or 200 coefficients in the filter. Additional coefficients will improve the response. Use a number four times higher.
Early bird savings extended! T4: Tools, Trends, Techniques, and Technology Monday, July 24, 2017 – Wednesday, July 26, 2017 Join BIOPAC for three days of hands-on, small-group human physiology workshops, presented by renowned experts, on the beautiful University of California at Santa Barbara campus in Santa Barbara, California, USA. Who should attend Anyone who records and […]
BIOPAC provides software and hardware that allow for research teams to record and analyze respiration activity in physiological experimentation. Here are a few notable studies in monitoring respiration and cardiovascular activity. The coupling between peripheral microcirculation and slow breathing: The purpose of this study was to investigate the coupling of breathing movements and microcirculatory blood […]
CBS News | 60 Minutes | Brain Hacking APRIL 9, 2017| Why can’t we stop looking at our smartphones? And are the designers of the apps and content on them using brain science to keep us hooked? Anderson Cooper reports. BIOPAC users Larry D. Rosen, Ph.D. and Nancy A. Cheever, Ph.D. from California State […]