The best solution is to avoid the ECG artifact in the first place. The problem is typically solved by adjusting the placement of the electrodes. Place the EMG electrodes so that they are NOT aligned with the axis of electrical activity of the heart. Placing both EMG electrodes (Vin+ and Vin- ) on the same side of the body will help.
ECG artifact can be removed from EMG data that has been recorded by applying a 50Hz High pass FIR filter. The number of coefficients should be greater than or equal to two times the sampling rate, divided by the lowest cutoff frequency specified. For example, if running a low pass filter at 1Hz on data sampled at 100Hz, choose at least (2 X 100/1) or 200 coefficients in the filter. Additional coefficients will improve the response. Use a number four times higher.
Smart shirt simultaneously acquires ECG and Respiration data while subjects roam freely The lightweight, comfortable BioNomadix BioShirt contains a respiration sensor and fabric electrodes to wirelessly record both respiration and ECG while ambulatory subjects move freely and perform tasks in short or long-term studies, in the lab, or in the real world. The shirt connects to a wireless BioNomadix BN-RSPEC Respiration […]
Research posters presented at BIOPAC’s T4 physiology conference (Tools, Trends, Techniques, and Technology) exhibited leading physiological research utilizing BIOPAC’s technology. Research posters presented unique advances in neuromarketing and health research. The T4 conference, held at UCSB Monday, July 24, 2017 through Wednesday, July 26, 2017, showcased the latest trends in the Tech industry’s physiological research. […]
ABC News | Good Morning America | Excessive Cell Phone Use JULY 31, 2017 | Excessive cell phone use may cause anxiety, experts warn BIOPAC user Nancy A. Cheever, Ph.D. from California State University, Dominguez Hills was featured on Good Morning America. Her study investigated whether spending too much time on your phone may be causing people to feel […]