The best solution is to avoid the ECG artifact in the first place. The problem is typically solved by adjusting the placement of the electrodes. Place the EMG electrodes so that they are NOT aligned with the axis of electrical activity of the heart. Placing both EMG electrodes (Vin+ and Vin- ) on the same side of the body will help.
ECG artifact can be removed from EMG data that has been recorded by applying a 50Hz High pass FIR filter. The number of coefficients should be greater than or equal to two times the sampling rate, divided by the lowest cutoff frequency specified. For example, if running a low pass filter at 1Hz on data sampled at 100Hz, choose at least (2 X 100/1) or 200 coefficients in the filter. Additional coefficients will improve the response. Use a number four times higher.
BSL Home rentals enable students to conduct hands-on physiology labs in the safety of their own home, dorm room, or any remote location while social distancing and stay-at-home orders keep them out of campus labs Complete lab solution to preserve practical lab experience! For instructors—BSL Home makes it easy for instructors to continue running practical […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]