It is possible to validate the signals coming from any DA100A/B/ or C amplifier by using a CBLCAL* Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ± 10 V range. Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCAL is connected to. The CBLCAL will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different than the source signal level, contact BIOPAC for further assistance.
* Note: Most other amplifiers require CBLCALC, not CBLCAL.
This training shows attendees how to create simple workflows, from opening files, to conditioning and preparing data for analysis, to using AcqKnowledge automated analysis tools for the extraction of meaningful information.
What Is Covered: – Opening graph files and displaying data – Channel Management – Transformations – Standardized conditioning protocols for analysis preparation – Analysis routines and custom sequences – Customized Batch Processing routines for your study protocol and more…
BIOPAC tools are used for studies in a variety of different fields and disciplines. AcqKnowledge software is one of our universally applicable tools, used by thousands of labs around the world and cited in thousands of peer reviewed publications, such as the following studies… The psychophysiology of guilt in healthy adults. Cognitive, Affective, & Behavioral Neuroscience, 1-18. […]
We had the opportunity to meet lots of researchers at our 4T|Phys Human Physiology Conference this month and demonstrate solutions for life science studies. We presented 39 unique skills sessions, including sessions on measuring blood pressure, recording HRV, and utilizing VR. These studies all apply these skills and further demonstrate how to apply our equipment… Real-Time Personalized Physiologically […]