It is possible to validate the signals coming from any DA100A/B/ or C amplifier by using a CBLCAL* Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ± 10 V range. Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCAL is connected to. The CBLCAL will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different than the source signal level, contact BIOPAC for further assistance.
* Note: Most other amplifiers require CBLCALC, not CBLCAL.
Learn how physiological signals from the body can help researchers and industrial designers improve ergonomics and reduce operator fatigue.
You Will Learn: – How to measure fatigue and ergonomic performance – Physiological signals used for ergonomics and fatigue – How to identify elevated or reduced stress – How to combine tools to better understand ergonomic conditions – How to rapidly prototype layouts using VR for improved ergonomics Watch On-Demand
Whether using specialized equipment, performing repetitive tasks, or simply going about our daily work routine, the effects of design in our environment as well as the tools we use and the way we use them can impact our health, work performance, and quality of life. Here are a few studies that have recorded and analyzed […]