It is possible to validate the signals coming from any DA100A/B/ or C amplifier by using a CBLCAL* Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ± 10 V range. Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCAL is connected to. The CBLCAL will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different than the source signal level, contact BIOPAC for further assistance.
* Note: Most other amplifiers require CBLCALC, not CBLCAL.
Add fNIRS Lessons to a C Series Imager Record NIRS measurements from the forehead and the forearm at two wavelengths, 730 nm for deoxy-Hb and 850 nm for oxy-Hb. Students embark on real-world NIRS research using a student-friendly approach in these four guided lessons covering nine experiments. COBI Modern makes NIRS experimentation easy for students […]
Researchers utilize electrodermal activity (EDA) data in a wide array of protocols. These recent studies join thousands of BIOPAC citations for EDA and represent just a few of BIOPAC’s hardware options for wired, wireless, logged, or MRI protocols with reusable or disposable EDA accessories and AcqKnowledge software solutions for Automated EDA Analysis Routines and EDA Measurement Tools. Caffeine Delivery […]