It is possible to validate the signals coming from any DA100A/B/ or C amplifier by using a CBLCAL* Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ± 10 V range. Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCAL is connected to. The CBLCAL will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different than the source signal level, contact BIOPAC for further assistance.
* Note: Most other amplifiers require CBLCALC, not CBLCAL.
T4: Tools, Trends, Techniques, and Technology Monday, July 24, 2017 – Wednesday, July 26, 2017 Join BIOPAC for three days of hands-on, small-group human physiology workshops, presented by renowned experts, on the beautiful University of California at Santa Barbara campus in Santa Barbara, California, USA. Who should attend Anyone who records and analyzes human physiology […]
The Human Anatomy & Physiology Society (HAPS) found that BIOPAC’s Student Lab System (BSL) and AcqKnowledge software were the number one ranked data acquisition and analysis systems for physiology experiments performed in introductory A&P labs. According to HAPS surveys conducted in 2013 and 2016, BIOPAC products are used and preferred by members in more labs than […]
Hardware and software packages from BIOPAC allow commercial consumer marketing teams to easily combine precise eye tracking data with physiological data recording. Here are a few notable studies recording physiology with eye tracking data: Predicting Advertising Success: While neurophysiological methods have become increasingly popular to better understand marketing phenomena among academics and practitioners, the success of […]