It is possible to validate the signals coming from any DA100A/B/ or C amplifier by using a CBLCAL* Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ± 10 V range. Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCAL is connected to. The CBLCAL will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different than the source signal level, contact BIOPAC for further assistance.
* Note: Most other amplifiers require CBLCALC, not CBLCAL.
The STMEPM Programmable Stimulation System for E-Prime allows a user to interface the STMISOLA Stimulator with E-Prime to control the stimulus frequency and stimulus intensity for real-time stimulus delivery changes based on a subject’s responses. It is also possible to hardcode the stimulus intensity levels in the presentation so that predefined stimulus levels are delivered […]
Academics and professionals in the field of human physiology gathered at UCSB for the inaugural T4 Physiology Conference. BIOPAC coordinated the three day conference held in late July, which attracted attendees from 26 different countries. The attendees gathered to learn the latest tools, trends, techniques and technology used in life science measurement and interpretation. The […]
Industry reporting firm medGadget recognized BIOPAC as a major market competitor in the global Blood Flow Measurement Devices market. BIOPAC hardware solutions for blood flow measurement include laser Doppler flowmetry and extensive online and offline analysis of raw blood flow data to take an array of calculations and measurements, such as Pulsatile max, Pulsatile Min, […]