It is possible to validate the signals coming from any DA100A/B/ or C amplifier by using a CBLCAL* Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ± 10 V range. Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCAL is connected to. The CBLCAL will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different than the source signal level, contact BIOPAC for further assistance.
* Note: Most other amplifiers require CBLCALC, not CBLCAL.
Smart Center Stand-alone, Portable Wireless Physiology Systems Smart Center Essentials entry-level system includes a Smart Center Device (BN-SMART with USB power cable), 2 or 3 Transmitters (BN-xxx-T with chargers); AcqKnowledge for Smart Center, and a case. Smart Center Enhanced adds a Logger (BN-LOGGER) and Basic Scripting License (ACK100W-BAS) to the Smart Center Device (BN-SMART with powercable), 3 […]
Science Night at Adams Elementary (Santa Barbara, CA) encourages students to discover, play, explore and learn about the importance of science and engineering in daily life. BIOPAC set up a BioNomadix Wireless Smart Center to allow students to record their heart rate. Electrodes were placed on students and leads were connected to a wireless transmitter that sent data […]
Recent developments regarding electric stimulation and mental processing have lead to interesting developments in psychophysiology and physiological sciences. The following studies look at developments in how we potentially care for and understand solutions to issues with mental health, as well as new ways to monitor electrical stimulus on participants. The ventromedial prefrontal cortex (vmPFC) is […]