It is possible to validate the signals coming from any DA100A/B/ or C amplifier by using a CBLCAL* Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ± 10 V range. Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCAL is connected to. The CBLCAL will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different than the source signal level, contact BIOPAC for further assistance.
* Note: Most other amplifiers require CBLCALC, not CBLCAL.
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]