It is possible to validate the signals coming from any DA100A/B/ or C amplifier by using a CBLCAL* Calibration Cable. The cable connects to the front of the amplifier and the UIM100C Analog Output 0 or 1, or to a known signal source in the ± 10 V range. Create a stimulus signal using AcqKnowledge and monitor the output of the amplifier that the CBLCAL is connected to. The CBLCAL will attenuate the output signal (1/1,000), and the amplifier will amplify the signal by the gain setting.
For example, if the stimulus signal is 1 V and the gain setting is 1,000, the corresponding signal from the amplifier should be 1 V. The signal can be used for amplifier validation and signal calibration.
If the signal is significantly different than the source signal level, contact BIOPAC for further assistance.
* Note: Most other amplifiers require CBLCALC, not CBLCAL.
Unity® Interface for AcqKnowledge® allows you to easily to connect your Unity3D projects with BIOPAC acquisition hardware and analysis software. Create your virtual environment using industry-standard Unity Connect and configure your project with AcqKnowledge in real time Control Acquisition from Unity to Custom Markers, Digital, and Analog I/O Deploy to your devices. Immerse your users and […]
Technology is more interactive in a multitude of contexts, from completing tasks at work, looking for help online, to leisurely activities like news and video streaming. Understanding how to improve interaction between humans and computers can hold benefits in multiple situations. The following are recent human-computer interaction (HCI) studies. The Eyes Have It Nasser et al. […]
With new developments happening daily, it’s easy to forget to slow down and take a breath. Breathing can be as forgettable as it is crucial; the often forgettable automatic function inspires rituals in religions around the world. Studies have shown the importance of breathing not just in a spiritual sense but also in physiological health benefits. Lung functionality […]