The TSD116C has nine wires that have to be connected to the screw terminals on the rear of the UIM100C module.
The black wire must be connected to the 5V terminal and the remaining eight wires must be connected to any of the digital input terminals. Loosen the screw on the terminal (for an MP150 system, the UIM100C must be separated from the MP150 side in order to expose the heads of the screws), insert the bare end of the wire, and tighten the screw. After all wires have been attached, re-connect the UIM100C to the rest of the system. Make sure that the wires are properly secured in the terminals before starting to use the switch.
High-performance mobile fNIR Imaging Systems are stand-alone functional brain imaging solutions for continuous NIR spectroscopy (NIRS). These top of the line systems provide cognitive function assessment and eliminate a great many of the drawbacks of a functional MRI. Mobile subjects are comfortable and can respond naturally in real-world situations to take a test or perform mobile […]
Recent Studies using BIOPACs MP-Series (MP160, MP150) data recorders, ECG Amplifiers, and BioNomadix Wireless recorders: Singing style to infants Mothers around the world sing to infants, presumably to regulate their mood and arousal. Lullabies and playsongs differ stylistically and have distinctive goals. Mothers sing lullabies to soothe and calm infants, and playsongs to engage and […]
Alex Dimov, BIOPAC Sales Executive, was interviewed at the Neuromarketing World Forum by Inside Marketing to discuss how virtual reality can improve marketing research. Alex explains how VR can be used to present virtual iterations of a product to save time and money, and notes that measuring physiological responses adds lots of information for product and messaging […]