The TSD116C has nine wires that have to be connected to the screw terminals on the rear of the UIM100C module.
The black wire must be connected to the 5V terminal and the remaining eight wires must be connected to any of the digital input terminals. Loosen the screw on the terminal (for an MP150 system, the UIM100C must be separated from the MP150 side in order to expose the heads of the screws), insert the bare end of the wire, and tighten the screw. After all wires have been attached, re-connect the UIM100C to the rest of the system. Make sure that the wires are properly secured in the terminals before starting to use the switch.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]