The TSD116C has nine wires that have to be connected to the screw terminals on the rear of the UIM100C module.
The black wire must be connected to the 5V terminal and the remaining eight wires must be connected to any of the digital input terminals. Loosen the screw on the terminal (for an MP150 system, the UIM100C must be separated from the MP150 side in order to expose the heads of the screws), insert the bare end of the wire, and tighten the screw. After all wires have been attached, re-connect the UIM100C to the rest of the system. Make sure that the wires are properly secured in the terminals before starting to use the switch.
The NICO Smart Amplifier is specifically designed to measure thoracic impedance for performing noninvasive cardiac output measurements. The amplifier terminates in two connectors for impedance (Z) and its derivative (dZ), and has a 3 m cable that connects directly to an AMI100D Amplifier Input Module. Use with EL500 series disposable electrodes and LEAD131 for a four-spot […]
NIRS and CNAP—Pairing great tools for great data Cerebral hemoglobin concentration (blood mass in the brain) is frequently measured in the prefrontal cortex, and systemic arterial blood pressure is known to be a variable factor based on the hemoglobin concentration. A leg pressure cuff was used to constrict and release blood flow to create a […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]