The TSD116C has nine wires that have to be connected to the screw terminals on the rear of the UIM100C module.
The black wire must be connected to the 5V terminal and the remaining eight wires must be connected to any of the digital input terminals. Loosen the screw on the terminal (for an MP150 system, the UIM100C must be separated from the MP150 side in order to expose the heads of the screws), insert the bare end of the wire, and tighten the screw. After all wires have been attached, re-connect the UIM100C to the rest of the system. Make sure that the wires are properly secured in the terminals before starting to use the switch.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Studying Cognitive Activity with fNIR! The age-old game of chess is known as a thinking man’s game, where foresight, critical thinking, and strategy are required to become a good chess player, and the terms “genius” or “brilliance” are bestowed upon the chess masters of the world. As such, chess is an excellent cognitive tool to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]