The TSD116C has nine wires that have to be connected to the screw terminals on the rear of the UIM100C module.
The black wire must be connected to the 5V terminal and the remaining eight wires must be connected to any of the digital input terminals. Loosen the screw on the terminal (for an MP150 system, the UIM100C must be separated from the MP150 side in order to expose the heads of the screws), insert the bare end of the wire, and tighten the screw. After all wires have been attached, re-connect the UIM100C to the rest of the system. Make sure that the wires are properly secured in the terminals before starting to use the switch.
Learn how to use AcqKnowledge to stimulate and record responses.
You will learn about: – Introduction to EPs – Theory and guidelines – How to record great EP data – How to record exogenous EPs (BAEP, VEP, SEP) – How to prepare data for analysis – Analysis automation
AcqKnowledge is an interactive, versatile software program that allows you to manage data in real time. Perform complex data acquisition, stimulation, triggering and analyses using intuitive navigation. Here are some recent AcqKnowledge citations… The nature of vicarious embarrassment. Journal of Business Research, 153, 355-364. Ziegler, A. H., Allen, A. M., Peloza, J., & Norris, J. I. (2022). […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]