AcqKnowledge 3.8.2 and above for Windows allows you to synchronize video files with physiological data. This allows you to jump to specific points in either the video or data file, and the corresponding data or video will jump to the correct time point. If you use a visual presentation system such as SuperLab, you can jump to the event markers in the data file and view the corresponding video.
Use an OUT103 LED cable to synchronize the video camera to the MP System. OUT103 connects to one of the analog output connectors on the bottom front UIM100C or the digital connectors on the back (for an MP150 system, the UIM will have to be separated at the MP150 side of the device to allow access to the heads of the screw terminals). To use OUT103 with an MP36R unit, add a BSLCBL6 interface cable and OUT3 adapter. In the software menu, under MP1XX>Setup Stimulator (in AcqKnowledge 4.4, choose “Stimulator” in the pane on the left after selecting “MP150 > Set Up Data Acquisition…”), use the stimulator settings to turn on the LED when the MP system starts recording data. Make sure that the video camera is recording and place the LED somewhere in view of the camera so that when you start recording you will see the LED turn on. When you review the video, you will see the LED turn on and that marks the start of the data acquisition. Use the timing measurements in AcqKnowledge to synchronize the video and data files.
The NICO Smart Amplifier is specifically designed to measure thoracic impedance for performing noninvasive cardiac output measurements. The amplifier terminates in two connectors for impedance (Z) and its derivative (dZ), and has a 3 m cable that connects directly to an AMI100D Amplifier Input Module. Use with EL500 series disposable electrodes and LEAD131 for a four-spot […]
NIRS and CNAP—Pairing great tools for great data Cerebral hemoglobin concentration (blood mass in the brain) is frequently measured in the prefrontal cortex, and systemic arterial blood pressure is known to be a variable factor based on the hemoglobin concentration. A leg pressure cuff was used to constrict and release blood flow to create a […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]