AcqKnowledge 3.8.2 and above for Windows allows you to synchronize video files with physiological data. This allows you to jump to specific points in either the video or data file, and the corresponding data or video will jump to the correct time point. If you use a visual presentation system such as SuperLab, you can jump to the event markers in the data file and view the corresponding video.
Use an OUT103 LED cable to synchronize the video camera to the MP System. OUT103 connects to one of the analog output connectors on the bottom front UIM100C or the digital connectors on the back (for an MP150 system, the UIM will have to be separated at the MP150 side of the device to allow access to the heads of the screw terminals). To use OUT103 with an MP36R unit, add a BSLCBL6 interface cable and OUT3 adapter. In the software menu, under MP1XX>Setup Stimulator (in AcqKnowledge 4.4, choose “Stimulator” in the pane on the left after selecting “MP150 > Set Up Data Acquisition…”), use the stimulator settings to turn on the LED when the MP system starts recording data. Make sure that the video camera is recording and place the LED somewhere in view of the camera so that when you start recording you will see the LED turn on. When you review the video, you will see the LED turn on and that marks the start of the data acquisition. Use the timing measurements in AcqKnowledge to synchronize the video and data files.
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
BIOPAC provides software and hardware that allows research teams to run autonomic function tests while collecting physiological data. Here are a few studies focusing on BIOPAC’s ability to record physiological signals during autonomic function testing. The Internet, Sleep, and Heart Rate Does internet usage lead to poorer sleep? Using a BIOPAC Data Acquisition System to measure cardiovascular […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]