Before performing a firmware upgrade, check compatibility. Make certain that no other users are connected to or using the MP150 you plan to upgrade.
Any attempts to communicate with the MP150 via any users AcqKnowledge software may affect the viability of the upgrade.
Fluctuations in the power source during an upgrade may also affect the outcome.
To use an MP150 with UDP communication on a network with a non-Windows DHCP server, you must use firmware 1.1.12 for the MP unit to properly be assigned an IP address. This is also true for any DHCP system with non-Windows operating systems, such as Unix, Linux, Mac OS X, and other DHCP-aware devices.
Launch AcqKnowledge 3.9.
Select MP150 > Update Firmware.
Click OK at the confirmation prompt.
Navigate to the MP150.mpb file and click Open.
The default location is C:Program Files BIOPAC Systems IncAcqKnowledge 3.9.
Click Start to upload the selected firmware file.
Wait for the confirmation prompt (usually less than 30 seconds) and then click OK.
The MP150 is delivered with a factory-programmed firmware version that is not affected by the upgrade procedure. You can always restore the original firmware with the Firmware Rollback Switch.
BSL Home rentals enable students to conduct hands-on physiology labs in the safety of their own home, dorm room, or any remote location while social distancing and stay-at-home orders keep them out of campus labs Complete lab solution to preserve practical lab experience! For instructors—BSL Home makes it easy for instructors to continue running practical […]
Why do we so easily believe Fake News? 68% of Americans find their news through social media, and the prevalence of incomplete or deceptive news stories dubbed “fake news” has wormed its way into the fabric of modern-day media. In an effort to study how social media based news is perceived, this study measured physiological […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]