PPG setup and calibration To reco rd a pulse waveform using the PPG100C, follow these setup and calibration details: Connections
Plug the transducer into the amplifier but do not connect it to the subject yet.
red lead into VSUP+ port
black lead into GND port
blue lead into INPUT port
Switch Settings : Using the 3.0 Hz LP setting and the 0.5 Hz HP setting in conjunction will make it easier to find the rate of the pulse waveform (these filters can be applied in software as well). Hint Gain: 10 (lowest Gain option) = top position LP: 3.0 Hz = top position HP (first bank) : 0.5 Hz HP = top position HP (second bank): 0.05 Hz HP = top position Set Up Acquisition
Recommended acquisition sample rate: 100 Hz.
Recording Start recording in Acq Knowledge. No calibration is necessary—simply make sure the correct channel is enabled and no other amplifiers are set to the same channel. Perform Amplifier baseline offset adjustment. After the amplifier baseline offset adjustment is complete, place the transducer on the finger (e.g., index finger tip) and tighten the Velcro strap (or attach to the ear with adhesive collars). Note: To minimize motion artifact, place the sensor on a hand that will remain mostly immobile. Check the tension level. If the strap is too tight, blood flow may be constricted, resulting in an almost straight line signal. If the strap is too loose, ambient light could enter the sensor, resulting in no recognizable waveform. Once you are able to obtain a good recording, it is recommended to increase the gain as much as possible as long as the pulse waveform never goes beyond ±10 Volts. The 0.5 Hz HP filter setting will force the signal to return to baseline faster than with the 0.05 Hz filter. The 10 Hz LP filter will reveal more detail from the waveform than the 3 Hz LP filter. Observe the effect of different filter settings (note that any additional filtering can also be performed in the software, either online or offline). Sample Results
The following graphs show results of motion artifact and varying tension levels:
Example of motion artifact (finger):
Example of movement artifact (hand):
Associated Applications Associated Application Notes
The BioNomadix two channel wireless goniometry amplifier is a dual output device that provide simultaneous measurements around two orthogonal rotational axes (e.g., wrist flexion/extension and radial/ulnar deviations). These units interface with the MP160 (or older MP150) data acquisition and analysis platform and AcqKnowledge software, allowing advanced analysis for multiple applications and supporting acquisition of a broad range of signals and measurements. BN-GONIO can support any […]
Whether using specialized equipment, performing repetitive tasks, or simply going about our daily work routine, the effects of design in our environment as well as the tools we use and the way we use them can impact our health, work performance, and quality of life. Here are a few studies that have recorded and analyzed […]
BIOPAC’s comprehensive EDA Guide provides an introduction to Electrodermal Activity (EDA or GSR) and details topics including: EDA Complex: SCL, SCR, tonic, phasic, specific SCR, non-specific SCR Participant Prep & Electrode Placement Data Recording tips Automated EDA Analysis Routines Digital input to Stim Events Stim-Response Analysis Derive Phasic EDA from Tonic Event-related EDA Analysis Locate […]