If you are using a single MP100/MP150 system, the maximum number of EEG amplifiers you can have is 16. If more channels are required, it is possible to link acquisitions from more than one MP150. However, the modular design of MP150 systems does not lend itself well to high density EEG recording, so BIOPAC does not strongly recommend this solution.
The Mobita System
32-channel wireless EEG system is a better solution for multi-channel applications.
For unipolar recording you need as many amplifiers as you have sites to record from. For example, for 16 sites you would need 16 EEG amplifiers. To perform this recording you would need 16 active electrodes, 1 reference (linked via 15 jumper cables to connect all amplifiers), and 1 ground.
For bipolar recording the number of EEG amplifiers will be the same as the number of pairs of electrode sites for which you want to calculate the potential difference. Therefore, if you want to look at all possible pairs, you can calculate the combinations without repetition: n!/(r!(n-r)!) where n is the number of electrode sites and r is 2. For example, for a bipolar recording on 5 sites you would need 5!/(2!3!) = 10 EEG amplifiers. This assumes that you want to look at the potential difference between all possible combinations of electrode sites. This is the maximum number of EEG amplifiers that would be required.
Here is a reference for the number of EEG amplifiers required if you need to look at all electrode pair potential differences:
2 electrodes: 1 amplifier
3 electrodes: 3 amplifiers
4 electrodes: 6 amplifiers
5 electrodes: 10 amplifiers
6 electrodes: 15 amplifiers
Page last modified 30Dec2014
Learn a technique—Trans-radial Electrical Bioimpedance Velocimetry (TREV)—that is more user-friendly and less invasive than the transthoracic method. TREV measures blood movement in the radial and ulnar arteries of the forearm via bioimpedance. Register Now
– Presentation and discussion of TREV
– Derived Indices Relevant to TREV
– Hardware setup
– Participant setup
– Recognizing good quality and bad quality data
– Demonstration with the grip anticipation test
– Review of results from other tests
Watch On Demand
Exercise physiology is the study of the body’s response to physical activity. Research in exercise physiology allows for insights in health and for a better understanding of the functions of the body at work. Here are some recent studies in exercise physiology that utilized BIOPAC tools for their research… Effect of exercise on electrocardiographic parameters […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […] Read All