If you are using a single MP100/MP150 system, the maximum number of EEG amplifiers you can have is 16. If more channels are required, it is possible to link acquisitions from more than one MP150. However, the modular design of MP150 systems does not lend itself well to high density EEG recording, so BIOPAC does not strongly recommend this solution.
The Mobita System 32-channel wireless EEG system is a better solution for multi-channel applications.
For unipolar recording you need as many amplifiers as you have sites to record from. For example, for 16 sites you would need 16 EEG amplifiers. To perform this recording you would need 16 active electrodes, 1 reference (linked via 15 jumper cables to connect all amplifiers), and 1 ground.
For bipolar recording the number of EEG amplifiers will be the same as the number of pairs of electrode sites for which you want to calculate the potential difference. Therefore, if you want to look at all possible pairs, you can calculate the combinations without repetition: n!/(r!(n-r)!) where n is the number of electrode sites and r is 2. For example, for a bipolar recording on 5 sites you would need 5!/(2!3!) = 10 EEG amplifiers. This assumes that you want to look at the potential difference between all possible combinations of electrode sites. This is the maximum number of EEG amplifiers that would be required.
Here is a reference for the number of EEG amplifiers required if you need to look at all electrode pair potential differences:
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
BIOPAC provides software and hardware that allows research teams to run autonomic function tests while collecting physiological data. Here are a few studies focusing on BIOPAC’s ability to record physiological signals during autonomic function testing. The Internet, Sleep, and Heart Rate Does internet usage lead to poorer sleep? Using a BIOPAC Data Acquisition System to measure cardiovascular […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]