The STP100C receives a digital pulse coming from a stimulus presentation program, either through the 37 pin connector or the 25 pin connector.
To route a digital pulse from a presentation program, for example to drive another device outside of the MP system, take a JUMP100 and insert into the Digital I/O port on the back of the UIM100C (note, the UIM100C is separated and the port strip has a screw to secure the JUMP100C).
Route the other end of the jumper cable into digital channels 0 – 3 on the UIM100C, this corresponds to the Coaxial outputs on the front of the STP100C 1-4 respectively.
Next route a JUMP100 from GRND D on the UIM100C to channel 4 digital I/O on the UIM100C. This completes the circuit and activates the front relays.
In the software, go to MP150>SHOW MANUAL CONTROL.
In the MANUAL CONTROL dialog box, set the DIGITAL I/O for Channels 0 -7 to INPUT. Set the DIGITAL I/O for channels 8 – 15 to read INPUT as well.
As an example, you have a digital TTL pulse coming into the back of the STP100C for channel 8. This channel will be enabled in AcqKnowledge and every time a pulse comes through to that channel, it will route to channel 0. A pulse will be seen coming out of BNC output #1 on the STP100C.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
MIT and Harvard researchers developed a new technique to expand ventilator capacity Individualized System for Augmenting Ventilator Efficacy (iSAVE): A Rapidly deployable system to expand ventilator capacity “The iSAVE is a rapidly deployable solution in response to the urgent shortage of mechanical ventilators for respiratory support during the COVID-19 pandemic. In contrast to previously described […]
New Ventilator Validation System Allows Device Manufacturers to Test Ventilator Pressure and Volume System supports rapid development and testing of hardware for COVID-19 treatment Goleta, California – March 24, 2020 BIOPAC Systems, Inc. announced a new Ventilator Validation System for engineering and device manufacturers to use when designing and testing new and existing ventilators for […]