The STP100C receives a digital pulse coming from a stimulus presentation program, either through the 37 pin connector or the 25 pin connector.
To route a digital pulse from a presentation program, for example to drive another device outside of the MP system, take a JUMP100 and insert into the Digital I/O port on the back of the UIM100C (note, the UIM100C is separated and the port strip has a screw to secure the JUMP100C).
Route the other end of the jumper cable into digital channels 0 – 3 on the UIM100C, this corresponds to the Coaxial outputs on the front of the STP100C 1-4 respectively.
Next route a JUMP100 from GRND D on the UIM100C to channel 4 digital I/O on the UIM100C. This completes the circuit and activates the front relays.
In the software, go to MP150>SHOW MANUAL CONTROL.
In the MANUAL CONTROL dialog box, set the DIGITAL I/O for Channels 0 -7 to INPUT. Set the DIGITAL I/O for channels 8 – 15 to read INPUT as well.
As an example, you have a digital TTL pulse coming into the back of the STP100C for channel 8. This channel will be enabled in AcqKnowledge and every time a pulse comes through to that channel, it will route to channel 0. A pulse will be seen coming out of BNC output #1 on the STP100C.
Continuous…Easy to use…Reproducible & Validated The NIBP100D-HD noninvasive blood pressure system provides a continuous, beat-to-beat, blood pressure signal recorded from the fingers of a subject. Simple and noninvasive—finger sensor provides accurate & immediate feedback: arterial BP, cardiac output, fluid and hemodynamic status. Measure noninvasive BP parameters (BP, sBP, dBP, mBP, and PR) plus hemodynamic parameters (PPV, […]
Alex Dimov, BIOPAC Sales Executive, was interviewed at the Neuromarketing World Forum by Inside Marketing to discuss how virtual reality can improve marketing research. Alex explains how VR can be used to present virtual iterations of a product to save time and money, and notes that measuring physiological responses adds lots of information for product and messaging […]
February 14, 2019 BIOPAC Systems, Inc. Goleta, CA USA New “Smart Amplifiers” make it easier than ever for life science researchers to get great data Fast Setup & High Performance Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level voltage connection to the data acquisition system and reduces noise artifact. […]