The STP100C receives a digital pulse coming from a stimulus presentation program, either through the 37 pin connector or the 25 pin connector.
To route a digital pulse from a presentation program, for example to drive another device outside of the MP system, take a JUMP100 and insert into the Digital I/O port on the back of the UIM100C (note, the UIM100C is separated and the port strip has a screw to secure the JUMP100C).
Route the other end of the jumper cable into digital channels 0 – 3 on the UIM100C, this corresponds to the Coaxial outputs on the front of the STP100C 1-4 respectively.
Next route a JUMP100 from GRND D on the UIM100C to channel 4 digital I/O on the UIM100C. This completes the circuit and activates the front relays.
In the software, go to MP150>SHOW MANUAL CONTROL.
In the MANUAL CONTROL dialog box, set the DIGITAL I/O for Channels 0 -7 to INPUT. Set the DIGITAL I/O for channels 8 – 15 to read INPUT as well.
As an example, you have a digital TTL pulse coming into the back of the STP100C for channel 8. This channel will be enabled in AcqKnowledge and every time a pulse comes through to that channel, it will route to channel 0. A pulse will be seen coming out of BNC output #1 on the STP100C.
HRV | Frequency Domain Measures You will learn about – Introduction to HRV – Theory and guidelines – Recording great ECG data – Preparing data for analysis – Low Frequency (LF) & High Frequency (HF) measures – Single-epoch spectral HRV analysis – Multi-epoch spectral HRV analysis – Automation with scripting and Workflow – Focus Areas and event-based analysis
BIOPAC provides researchers with a complete range of tools to gather data on heart rate variability (HRV). The following studies demonstrate just some of the ways in which HRV research benefits from the implementation of BIOPAC hardware and software solutions. PTSD and HRV Post-traumatic stress disorder (PTSD) has been associated with dysfunction of the autonomic […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]