The STP100C receives a digital pulse coming from a stimulus presentation program, either through the 37 pin connector or the 25 pin connector.
To route a digital pulse from a presentation program, for example to drive another device outside of the MP system, take a JUMP100 and insert into the Digital I/O port on the back of the UIM100C (note, the UIM100C is separated and the port strip has a screw to secure the JUMP100C).
Route the other end of the jumper cable into digital channels 0 – 3 on the UIM100C, this corresponds to the Coaxial outputs on the front of the STP100C 1-4 respectively.
Next route a JUMP100 from GRND D on the UIM100C to channel 4 digital I/O on the UIM100C. This completes the circuit and activates the front relays.
In the software, go to MP150>SHOW MANUAL CONTROL.
In the MANUAL CONTROL dialog box, set the DIGITAL I/O for Channels 0 -7 to INPUT. Set the DIGITAL I/O for channels 8 – 15 to read INPUT as well.
As an example, you have a digital TTL pulse coming into the back of the STP100C for channel 8. This channel will be enabled in AcqKnowledge and every time a pulse comes through to that channel, it will route to channel 0. A pulse will be seen coming out of BNC output #1 on the STP100C.
Workflow—A Great Tool for Every Researcher! Automate AcqKnowledge with New Workflow Tool New automation tool significantly saves time and improves consistency with standardized procedures Workflow’s drag-and-drop interface allows you to automate and repeat the signal-conditioning and processing steps your protocol requires in an easy-to-use process without programming or scripting knowledge. Batch process multiple files with confidence […]
Whether using specialized equipment, performing repetitive tasks, or simply going about our daily work routine, the effects of design in our environment as well as the tools we use and the way we use them can impact our health, work performance, and quality of life. Here are a few studies that have recorded and analyzed […]