How to eliminate artifacts caused by magnetic stimulation (do not confuse with MRI artifacts):
The magnetic stimulator could induce a current to flow in the (for example) EMG leads. A possible solution might be to twist the EMG recording leads in a tight spiral so the effects of the external magnetic field are minimized.
Another method would be to shield the leads using a highly magnetically conductive material (like permalloy). This material requires no electrical connection. A permalloy cable wrapped around the leads could help reduce artifact. Normal shielded leads would not help in this situation if a magnetic field looks to be the primary culprit. The shielded leads only work to minimize impacts of electric fields, not magnetic fields.
Finally, it may be possible to reduce or remove the noise in post-processing or real-time.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
Recent physiological studies feature BIOPAC’s Wireless BioNomadix Data Recorders Using Metronomes and Rhythmic stimulation to help people with Parkinson’s disease walk For those living with Parkinson’s disease, even simple tasks can take too long to accomplish properly, such as crossing a room or picking up an object. In this study, participants with diagnosed cases of […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]