How to eliminate artifacts caused by magnetic stimulation (do not confuse with MRI artifacts):
The magnetic stimulator could induce a current to flow in the (for example) EMG leads. A possible solution might be to twist the EMG recording leads in a tight spiral so the effects of the external magnetic field are minimized.
Another method would be to shield the leads using a highly magnetically conductive material (like permalloy). This material requires no electrical connection. A permalloy cable wrapped around the leads could help reduce artifact. Normal shielded leads would not help in this situation if a magnetic field looks to be the primary culprit. The shielded leads only work to minimize impacts of electric fields, not magnetic fields.
Finally, it may be possible to reduce or remove the noise in post-processing or real-time.
New Automation Tool Saves Time and Improves Data Consistency
Workflow’s drag-and-drop interface allows you to automate and repeat the signal-conditioning and processing steps your protocol requires in an easy-to-use process without programming or scripting. – Batch process multiple files with confidence that they’re all processed using the same procedure. – Expedite signal conditioning and preparation for analysis with a standardized and automated approach. – Extract measurements, analyze data and get results faster, for meaningful scientific insights. – Save and share workflows with others in your team to help reduce training time and increase collaboration.
AcqKnowledge is an interactive, versatile software program that allows you to manage data in real time. Perform complex data acquisition, stimulation, triggering and analyses using intuitive navigation. Here are some recent AcqKnowledge citations… The nature of vicarious embarrassment. Journal of Business Research, 153, 355-364. Ziegler, A. H., Allen, A. M., Peloza, J., & Norris, J. I. (2022). […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]