How to eliminate artifacts caused by magnetic stimulation (do not confuse with MRI artifacts):
The magnetic stimulator could induce a current to flow in the (for example) EMG leads. A possible solution might be to twist the EMG recording leads in a tight spiral so the effects of the external magnetic field are minimized.
Another method would be to shield the leads using a highly magnetically conductive material (like permalloy). This material requires no electrical connection. A permalloy cable wrapped around the leads could help reduce artifact. Normal shielded leads would not help in this situation if a magnetic field looks to be the primary culprit. The shielded leads only work to minimize impacts of electric fields, not magnetic fields.
Finally, it may be possible to reduce or remove the noise in post-processing or real-time.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
MIT and Harvard researchers developed a new technique to expand ventilator capacity Individualized System for Augmenting Ventilator Efficacy (iSAVE): A Rapidly deployable system to expand ventilator capacity “The iSAVE is a rapidly deployable solution in response to the urgent shortage of mechanical ventilators for respiratory support during the COVID-19 pandemic. In contrast to previously described […]
New Ventilator Validation System Allows Device Manufacturers to Test Ventilator Pressure and Volume System supports rapid development and testing of hardware for COVID-19 treatment Goleta, California – March 24, 2020 BIOPAC Systems, Inc. announced a new Ventilator Validation System for engineering and device manufacturers to use when designing and testing new and existing ventilators for […]