How to eliminate artifacts caused by magnetic stimulation (do not confuse with MRI artifacts):
The magnetic stimulator could induce a current to flow in the (for example) EMG leads. A possible solution might be to twist the EMG recording leads in a tight spiral so the effects of the external magnetic field are minimized.
Another method would be to shield the leads using a highly magnetically conductive material (like permalloy). This material requires no electrical connection. A permalloy cable wrapped around the leads could help reduce artifact. Normal shielded leads would not help in this situation if a magnetic field looks to be the primary culprit. The shielded leads only work to minimize impacts of electric fields, not magnetic fields.
Finally, it may be possible to reduce or remove the noise in post-processing or real-time.
High-performance mobile fNIR Imaging Systems are stand-alone functional brain imaging solutions for continuous NIR spectroscopy (NIRS). These top of the line systems provide cognitive function assessment and eliminate a great many of the drawbacks of a functional MRI. Mobile subjects are comfortable and can respond naturally in real-world situations to take a test or perform mobile […]
BIOPAC provides software and hardware that allows research teams to run autonomic function tests while collecting physiological data. Here are a few studies focusing on BIOPAC’s ability to record physiological signals during autonomic function testing. The Internet, Sleep, and Heart Rate Does internet usage lead to poorer sleep? Using a BIOPAC Data Acquisition System to measure cardiovascular […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]