In some MRI installations, there may be an offset in EDA measurements. The following steps can be performed in order to calibrate skin conductance measurements:
Install all cabling and place the EDA leads in the scanner, where they will be once the participant is there. The connection should be as follows: EDA leads->MRI cabling in chamber room to filter->Filter->MRI cabling from filter in control room->GSR100C amplifier.
The procedure involves performing a 2-point calibration with 2 known conductance values. You can record the voltage corresponding to conductance 0 (open leads) and conductance 10uMho (via placing a 100kOhm resistor between the leads) to calibrate the GSR100C amplifier. The GSR amplifier must be in DC mode (both high-pass filter switches should be set to DC). The following steps will guide you through the procedure.
MP menu->Set up channels and click on Setup for the GSR channel. Make sure the leads are open (nothing attached to the leads). Click on CAL1 and enter 0 for the Map value box for CAL1.
Attach a 100kOhm resistor between the leads and click on CAL2, enter 10 for the Map box for CAL2.
Set the units to uMho.
The amplifier will be calibrated for measurements in the scanner.
IMPORTANT! See Safety Guidelinesfor recording biopotential measurements in the MRI environment.
Eyes are the window to the soul and, more importantly, the brain. Psychophysiology, neuroscience, and consumer neuroscience researchers use eye tracking technology to understand emotion, behavior, subject response, decision-making, and human performance—and to help improve products and services. But not everyone knows where to start or how and why to use this important technology. A panel of eye tracking experts will present typical use cases and the latest eye tracking technology.
Join this online presentation with Q&A to learn the fundamentals of eye tracking!
Recent physiological studies feature BIOPAC’s Wireless BioNomadix Data Recorders Using Metronomes and Rhythmic stimulation to help people with Parkinson’s disease walk For those living with Parkinson’s disease, even simple tasks can take too long to accomplish properly, such as crossing a room or picking up an object. In this study, participants with diagnosed cases of […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]