In some MRI installations, there may be an offset in EDA measurements. The following steps can be performed in order to calibrate skin conductance measurements:
Install all cabling and place the EDA leads in the scanner, where they will be once the participant is there. The connection should be as follows: EDA leads->MRI cabling in chamber room to filter->Filter->MRI cabling from filter in control room->GSR100C amplifier.
The procedure involves performing a 2-point calibration with 2 known conductance values. You can record the voltage corresponding to conductance 0 (open leads) and conductance 10uMho (via placing a 100kOhm resistor between the leads) to calibrate the GSR100C amplifier. The GSR amplifier must be in DC mode (both high-pass filter switches should be set to DC). The following steps will guide you through the procedure.
MP menu->Set up channels and click on Setup for the GSR channel. Make sure the leads are open (nothing attached to the leads). Click on CAL1 and enter 0 for the Map value box for CAL1.
Attach a 100kOhm resistor between the leads and click on CAL2, enter 10 for the Map box for CAL2.
Set the units to uMho.
The amplifier will be calibrated for measurements in the scanner.
IMPORTANT! See Safety Guidelinesfor recording biopotential measurements in the MRI environment.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]