To record bipolar EEG, you will need an EEG amplifier for each pair of electrodes that you want to calculate the potential difference between. It is important to note that you can also derive bipolar measurements from unipolar measurements. This is done in software. For example, if you are recording from sites A, B, C in unipolar you can then get the following bipolar measurements:
AB by calculating A-B
BC by calculating B-C
AC by calculating A-C
If you want to look at all possible pairs, you can calculate the combinations without repetition: n!/(r!(n-r)!) where n is the number of electrodes and r is 2.
For example, for a bipolar recording on 5 sites you would need 5!/(2!3!) = 10 EEG amplifiers. This assumes that you want to look at the potential difference between all possible combinations of electrode sites. This is the maximum number of EEG amplifiers that would be required.
Here is a reference for the number of EEG amplifiers required if you need to look at all electrode pair potential differences:
The BioNomadix two channel wireless goniometry amplifier is a dual output device that provide simultaneous measurements around two orthogonal rotational axes (e.g., wrist flexion/extension and radial/ulnar deviations). These units interface with the MP160 (or older MP150) data acquisition and analysis platform and AcqKnowledge software, allowing advanced analysis for multiple applications and supporting acquisition of a broad range of signals and measurements. BN-GONIO can support any […]
Whether using specialized equipment, performing repetitive tasks, or simply going about our daily work routine, the effects of design in our environment as well as the tools we use and the way we use them can impact our health, work performance, and quality of life. Here are a few studies that have recorded and analyzed […]