In some instances, iLok drivers will not install properly or will fail to install, leading to connection failure between AcqKnowledge and the iLok USB Key.
If the iLok USB drivers did not succesfully install:
The driver executable will need to be activated either using the original installation disk, or by contacting BIOPAC support for driver download information.
Determine whether the computer is running a 32-bit or 64-bit operating system (right-click on the computer’s desktop icon or on “Computer” in the Windows “Start” menu and select “Properties” to find out)
On the installation disk, open the “supportcomponents” folder. Inside that folder, find the appropriate file: “ilokxx**setup.exe” (where “**” should be “32” or “64” depending upon the type of operating system). Double-click the file to run the program and follow the prompts.
If the iLok USB drivers installed but AcqKnowledge is unable to start with the USB key plugged in:
Open the “Programs and Features” Windows Control Panel, find the “ilokxx driver” (probably named “iLok Installer x32” or “iLok Installer x64”), and uninstall.
Re-install the driver as indicated in the steps above.
If the steps above do not solve the problem:
Unplug the iLok key and insert it into another USB port on the computer.
If the iLok is not operating properly
If you have the iLok key inserted, but the key is not being recognized even though the program is open, then please open files by choosing “File > Open…” in the menu of an existing AcqKnowledge graph window. If you try to open a file through an Explorer window when AcqKnowledge is already running, the operating system attempts to launch a second instance of AcqKnowledge. It will then try to authorize that instance but will fail because the iLok key is already being used by the first instance.
Learn how to maximize your BIOPAC data acquisition system and implement best practices for setting up and optimizing your MP System.
The presentation will cover the following: • Demonstrate complete hardware and software setup • Include details on how to incorporate commonly used amplifiers in the D-series, C-series, and BioNomadix • Review legacy issues related to older system components and workarounds • Show how to properly set up participants including proper lead options, electrode options, placement, and grounding/isolation procedures
BIOPAC provides software and hardware that allows research teams to study any physiological measure. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and HRV Analysis. Intelligent Interaction Leading to Solutions The research being done in the HCI space is amazing, as scientists continue to make computers more adept at communicating with humans, allowing more […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]