Transcranial Direct Current Stimulation (tDCS) is an electrical stimulation method, which employs a direct current of value typically between 1 and 2 milliamps (ma). This method is typically oriented to neuronal stimulation in the brain. Depending on the type of electrodes used for stimulation, and their associated placement on the scalp and body, Transcranial Direct Current Stimulation (tDCS) may also be referred to as High Definition Transcranial Direct Current Stimulation (HD-tDCS).
Upon first introduction tDCS employed the use of large surface area electrodes (25 cm2 – 64 cm2), and electrodes—either carbon composition or metal—were isolated from the skin surface with conductive gel soaked sponges, to help reduce ion transfer from electrodes into the skin and to reduce current density at the electrode sites. tDCS can also be employed with Ag/AgCl electrode arrays.
Using BIOPAC for tDCS and HD-tDCS…
Total arbitrary control of current wave shape is possible (DC or AC stimulation).
Multiple STMISOLA units can be applied to the same subject.
Electrode configurations are highly mutable.
Exact output current applied can be sampled.
Other physiological signals can be sampled simultaneously to look for tDCS impact on hemodynamics, sympathetic or parasympathetic nervous system.
This application note illustrates how a HD-tDCS setup can be constructed using the following BIOPAC equipment:
Sample tDCS Hardware and Setup
Basic setup for stimulation only:
1 x MP160/HLT or MP150/UIM (Data acquisition system) 1 x STMISOLA (Isolated linear stimulator) 1 x CBLCFMA (Current feedback monitor) 1 x CBLLIMIT2 (Current Limiter 2 ma) 3 x CBL204 (Y–electrode lead coupler) 5 x LEAD110 (Electrode lead, unshielded) 1 pk. EL502 or EL504 (Solid gel electrodes)
If also recording biopotentals at the same time:
MP160 setup: add 1 x INISO (Analog signal isolator)
MP150 setup: add 1 x HLT100C (High-level transducer amp) + 1 x INISO (Analog signal isolator)
Basic principles of physiological data collection covers fundamentals of data collection and equipment configuration equipment to help you refine experiment protocols and avoid costly missteps. Frazer Findlay, CEO of BIOPAC, discusses common mistakes with physiology recording and shares his secrets for collecting great data. Topics include: How to prep a subject and where to place electrodes; Which type of electrodes work for different body signals; Analog to Digital Conversion; Sample rates (correct vs. incorrect); Filtering; and Scaling and calibration.Watch On Demand Now!
Recent psychophysiological studies feature BIOPAC’s MP Research series of Data Recorders and AcqKnowledge software. Creating Adaptive Assistants for helicopter crews based on Real-time Physiological Data In the Mercedes S-Class, biometric tools track a driver’s eyes, heart rate, and other physiological data, with one goal in mind: Making sure drivers are alert and safe on the […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]