This Application Note describes how to record percent End-Tidal CO2 (% ETCO2) using a BIOPAC CO2100C Carbon Dioxide Amplifier. The note details subject interface options for high flow without a mask, high flow with a mask, low flow without a mask, and MRI.
When a BIOPAC CO2100C Carbon Dioxide Measurement Amplifier is used in conjunction with the MP150 Data Acquisition System and AcqKnowledge software, it’s easy to perform, real-time, breath-by-breath measurements. The AcqKnowledge software will calculate the peak expired CO2 signal to report percent End-Tidal CO2.
End-Tidal CO2 can also be combined with other pulmonary function measurements when a flow transducer and oxygen measurement amplifier are added to the system.
Workflow—A Great Tool for Every Researcher! Automate AcqKnowledge with New Workflow Tool New automation tool significantly saves time and improves consistency with standardized procedures Workflow’s drag-and-drop interface allows you to automate and repeat the signal-conditioning and processing steps your protocol requires in an easy-to-use process without programming or scripting knowledge. Batch process multiple files with confidence […]
AcqKnowledge is an interactive, versatile software program that allows you to manage data in real time. Perform complex data acquisition, stimulation, triggering and analyses using intuitive navigation. Here are some recent AcqKnowledge citations… The nature of vicarious embarrassment. Journal of Business Research, 153, 355-364. Ziegler, A. H., Allen, A. M., Peloza, J., & Norris, J. I. (2022). […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]