An electrocardiogram (ECG) is a graphical recording of the changes occurring in the electrical potentials between different sites on the skin as a result of cardiac activity. The electrical activity of the heart is a sequence of depolarizations and repolarizations. Depolarization occurs when the cardiac cells, which are electrically polarized, lose their internal negativity. The spread of depolarization travels from cell to cell, producing a wave of depolarization across the entire heart. This wave represents a flow of electricity that can be detected by electrodes placed on the surface of the body. Once depolarization is complete, the cardiac cells are restored to their resting potential, a process called repolarization.
This flow of energy takes on the form of the ECG wave and is characterized by an initial P wave, followed by the QRS complex, and then the T wave. The P wave is associated with depolarization of the atria, the QRS complex is associated with depolarization of the ventricles, and the T wave with repolarization of the ventricles.
This application note is designed to assist with the advanced analysis of ECG data, specifically P, Q, R, S and T Wave analysis. This Application Note describes how to set up the AcqKnowledge software to find R time, R height, Rate, Inter-Beat Interval, S height, P height, Q height, and T height and how to place your measurements into the journal.
ICG: Impedance Cardiography/Cardiac Output - Record cardiac output, thoracic impedance changes, or any kind of bioimpedance signal. Use AcqKnowledge for a fully automated ICG analysis and dZ/dt waveform classifier.
Sleep Studies - Long term recordings with up to 16 channels of data. Record EEG, EOG, EMG, respiration, temp., sound, limb position and more. Filter out EEG frequencies to score sleep stages.
ECG: Cardiology - Connect up to 16 ECG input leads with MP160 system. Use automated analysis features for HRV, RSA, to classify heartbeats, identify arrhythmias, and perform ECG averaging.
Cardiovascular Hemodynamics - Hardware for human and animals. Measure continuous BP, ECG, stroke volume, cardiac output, PPG, etc. For animal and tissue look at acute and chronic dose-response.
In vitro Pharmacology - Record from tissue bath, isolated organ, Ussing chamber, or use field stimulation. Advanced algorithms analyze Langendorff, working heart, and isolated perfused lung data.
Exercise Physiology - Examine ventilation, oxygen uptake, carbon dioxide production, biopotentials, temp., and biomechanical signals simultaneously. Record wirelessly with BioNomadix.
Amplifiers & Interfaces - Use BIOPAC amplifiers with MP Systems, as stand-alone devices, or with 3rd-party flow meters, force plates, sono-micrometers, telemetry equipment, metabolic carts, etc.
Stim-Response | Virtual Reality - Synchronize events from a virtual world with physiological data from an MP150 system. Use feedback loops for greater control and automation—change the VR world in real time.
The Ventilator Validation Kit supports new prototype development, existing design validation for production testing, mobile field testing and verification, and other uses where accurate pressure and flow measurements are critical. The Ventilator Validation Kit provides ventilator manufacturers and prototype developers with a complete solution for validating new medical/hospital-grade ventilator products intended for human use. The VVK100-SYS includes […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]
Comparing Muscle Movement and Involvement between Boxing Punches In the fast-paced, high-intensity combat sport of boxing, the right punch can win a fight. However, boxing is also a sport where opponents can become exhausted in mere minutes of action, and prioritizing their strength is key to winning a fight. This team recorded EMG on test […]