The most common form of indirect blood pressure measurement employs a pressure cuff, pump and pressure transducer. This complete assembly is commonly referred to as a Sphygmomanometer.
Typically, the cuff is wrapped around the upper arm and is inflated to a pressure exceeding that of the brachial artery. This amount of pressure collapses the artery and stops the flow of blood to the arm. The pressure of the cuff is slowly reduced as the pressure in the cuff is monitored by the pressure transducer. As the pressure drops, it will eventually match the systolic (peak) arterial pressure. At this point, the blood is able to “squirt” through the brachial artery. This squirting results in turbulence which creates the Korotkoff sounds. The Korotkoff sounds are detected using an SS17L physiological sounds transducer. As the cuff pressure continues to drop, the pressure will eventually match the diastolic pressure of the artery. At this point the Korotkoff sounds stop completely, because the blood is now flowing unrestricted through the artery.
Psychophysiology - Record and analyze BP, ECG, HRV, EDA, EMG, EEG, EOG, RSP, etc. Interface to stimulus presentation programs...use automated analysis routines to easily score and analyze data.
Exercise Physiology - Examine ventilation, oxygen uptake, carbon dioxide production, biopotentials, temp., and biomechanical signals simultaneously. Record wirelessly with BioNomadix.
Remote Monitoring - BioNomadix modules provide high quality, full-bandwidth data for a variety of signals—ECG, EEG, EGG, EMG, EOG, PPG, RSP, SKT, Accelerometry, Cardiac Output, Dynamometry, and Heel-Toe Strike.
Cardiovascular insights are much easier to achieve with this new integration from BIOPAC and CNSystems. The CNAP™ technology is now fully incorporated in a single BIOPAC amplifier for the MP160 system where it can be fully controlled by AcqKnowledge Software—optimized for research utilizing core CNAP clinical technology.
What you will learn: • Correct equipment set up and operation • Principles of the CNAP measurement technology • How to get reproducible data • How to interface equipment and optimize multiple measurements • Automatic blood pressure analysis • Application of the “Focus Areas” feature in AcqKnowledge Software
BIOPAC provides software and hardware that allows research teams to study physiological measures related to stress. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and EEG/ERP. Moms Moderating Stress Can relaxation techniques help lower stress on both mother and fetus? Using an MP36R Data Acquisition system to record ECG and EDA, this study […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]