This Application Note details setup and measurements for Auditory Brainstem Response (ABR) measurements using an MP160/MP150 data acquisition system and AcqKnowledge 4.1 or above software.
Whenever a sound wave is detected by the ear, it triggers a number of neuro-physiological responses along the auditory pathway. An action potential is conducted along the eighth nerve, the brainstem, and finally to the brain.
About one third of a second after initial stimulation, the signal evokes a response in the area of the brain where sounds are interpreted. These response signals are so small that they are frequently masked by the “background noise” of electrical activity in the brain. This interference is random with respect to the initial stimulation.
If you find an average of this interference signal, it will approach zero. However, since the evoked response is dependent on the stimulus, it should occur at the same time relative to the stimulus when the process is repeated. Therefore, if you repeat this stimulus-response process many times the “background noise” will disappear and what will be left is the neuro-physiological response of principal interest.
The ABR Calibration Guide outlines precise test equipment calibration to ensure proper signal-to-noise levels (S/N) and signal time delays for Auditory Brainstem Response (ABR) measurements.
For ABR studies, calibrate the OUT101 Tubephone with anEtymotic ER-7C Probe Microphone—this microphone provides a calibrated output voltage which is a function of applied Sound Pressure Level (SPL). The sensitivity is 50 mV/Pascal (-46 dB re: 1 V/uBar): 0 dB SPL = 0 dBuV. Place the Probe Microphone insert tube in the auditory canal prior to the insertion of the OUT101 foam tip. The OUT101 Tubephone sound delivery tube and the Probe Microphone sound input tube will then be exposed to the same auditory chamber. Accordingly, the SPL is recorded, via the Probe Microphone, simultaneously with applied auditory stimulus from the OUT101 Tubephone.
EEG: Electroencephalography - Wired and wireless EEG solutions. Record up to 32 channels of EEG and use software features for filtering, removing EOG artifacts, and complete frequency analysis.
ERS: Evoked Response - Record evoked, late, and field potentials, as well as startle and nerve conduction. Powerful on- and off-line averaging features (P300), and dual stimulation ability with the MP150.
Psychophysiology - Record and analyze BP, ECG, HRV, EDA, EMG, EEG, EOG, RSP, etc. Interface to stimulus presentation programs...use automated analysis routines to easily score and analyze data.
Amplifiers & Interfaces - Use BIOPAC amplifiers with MP Systems, as stand-alone devices, or with 3rd-party flow meters, force plates, sono-micrometers, telemetry equipment, metabolic carts, etc.
Smart shirt simultaneously acquires ECG and Respiration data while subjects roam freely The lightweight, comfortable BioNomadix BioShirt contains a respiration sensor and fabric electrodes to wirelessly record both respiration and ECG while ambulatory subjects move freely and perform tasks in short or long-term studies, in the lab, or in the real world. The shirt connects to a wireless BioNomadix BN-RSPEC Respiration […]
Research posters presented at BIOPAC’s T4 physiology conference (Tools, Trends, Techniques, and Technology) exhibited leading physiological research utilizing BIOPAC’s technology. Research posters presented unique advances in neuromarketing and health research. The T4 conference, held at UCSB Monday, July 24, 2017 through Wednesday, July 26, 2017, showcased the latest trends in the Tech industry’s physiological research. […]
ABC News | Good Morning America | Excessive Cell Phone Use JULY 31, 2017 | Excessive cell phone use may cause anxiety, experts warn BIOPAC user Nancy A. Cheever, Ph.D. from California State University, Dominguez Hills was featured on Good Morning America. Her study investigated whether spending too much time on your phone may be causing people to feel […]