This Application Note details various protocols for identifying a stimulation threshold for stimulation. The purpose of establishing this threshold is to ensure the subject’s comfort level and safely determine the point at which a stimulation signal becomes unpleasant, or even painful. This is accomplished by sending a series of very short stimulus pulses to the subject and gradually increasing the voltage. The subject then indicates his or her level of comfort or discomfort, and the stimulation is adjusted accordingly.
Case 1: MP150 + UIM100C + STM100C + STMISOLA (current and voltage mode) or STMISOC/D/E (voltage mode only) + LEAD110A x2 + EL509 + GEL104 Case 1a: MP160 + HLT100C + STM100C + STMISLOA or STMISOC/D/E + LEAD110A x2 + EL509 + GEL104
Case 2a: MP150 + UIM100C + STM200 + LEAD110A x2 + EL509 + GEL104 Case 2b: 3rd-party hardware and software + STM200 + LEAD110A x2 + EL509 + GEL104
Case 3: E-Prime + STMISOLA or STMISOC + LEAD110A x2 + EL509 + GEL104
ERS: Evoked Response - Record evoked, late, and field potentials, as well as startle and nerve conduction. Powerful on- and off-line averaging features (P300), and dual stimulation ability with the MP150.
Psychophysiology - Record and analyze BP, ECG, HRV, EDA, EMG, EEG, EOG, RSP, etc. Interface to stimulus presentation programs...use automated analysis routines to easily score and analyze data.
Stim-Response | Virtual Reality - Synchronize events from a virtual world with physiological data from an MP150 system. Use feedback loops for greater control and automation—change the VR world in real time.
Register Today! T4: Tools, Trends, Techniques, and Technology Monday, July 24, 2017 – Wednesday, July 26, 2017 Join BIOPAC for three days of hands-on, small-group human physiology workshops, presented by renowned experts, on the beautiful University of California at Santa Barbara campus in Santa Barbara, California, USA. Who should attend Anyone who records and analyzes […]
The Mobita 32-Channel EEG recording system was used in research examining if physiological data can be used to predict truly random events that correspond to perceptual stimuli. Baumgart, et al. from the University of California Santa Barbara utilized a quantum random number generator (qRNG) to choose from three randomized conditions: light, sound, and no stimulus. […]
BIOPAC provides software and hardware that allow for research teams to record and analyze respiration activity in physiological experimentation. Here are a few notable studies in monitoring respiration and cardiovascular activity. The coupling between peripheral microcirculation and slow breathing: The purpose of this study was to investigate the coupling of breathing movements and microcirculatory blood […]