If the positive and negative lead wires were merely swapped, then the signal will just be inverted (not otherwise distorted). The signal may be flipped over either with “Transform > Rescale…”:
or “Transform > Waveform Math…”:
In the case of “Rescale”, just select the inverted channel before choosing “Rescale”. In the case of “Waveform Math”, the source channel is specified through the dialog box, so it is not necessary to select the channel first.
The NIBP100E noninvasive blood pressure system provides continuous, beat‐to‐beat, blood pressure signal and values (Sys, Dia, MAP) and Pulse Rate (PR)*. The system outputs a continuous blood pressure waveform that is similar to a direct arterial pressure waveform. The amplifier provides values for systolic, diastolic, mean blood pressure, and heart rate. The NIBP100E amplifier with CNAP™ […]
BIOPAC provides software and hardware that allows research teams to study any physiological measure. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and HRV Analysis. Intelligent Interaction Leading to Solutions The research being done in the HCI space is amazing, as scientists continue to make computers more adept at communicating with humans, allowing more […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]