We do not recommend using the pulse transducer on the same hand as the NIBP100A transducer if you are performing pulse transit time measurements. The NIBP100A will have some influence on the PPG data because it applies a cyclic pressure (cycles every ~10-15 seconds) to the radial artery. During times of compression the pulse transit time will slow somewhat and the pulse signal will flatten out because the radial compression will act like a low-pass filter on the vasculature.
A work-around might be to use the NIBP100A and PPG on different hands.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
BIOPAC provides software and hardware that allows engineers to study the body. Here are a few notable studies using BIOPAC equipment for Biomedical Engineering research. Rehabilitating the Rehab Industry Could physical therapy be done from the comforts of your own home? If possible, this would open the door for more people with all sorts of ailments to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]