We do not recommend using the pulse transducer on the same hand as the NIBP100A transducer if you are performing pulse transit time measurements. The NIBP100A will have some influence on the PPG data because it applies a cyclic pressure (cycles every ~10-15 seconds) to the radial artery. During times of compression the pulse transit time will slow somewhat and the pulse signal will flatten out because the radial compression will act like a low-pass filter on the vasculature.
A work-around might be to use the NIBP100A and PPG on different hands.
MedelOpt® Mobility systems provide high-density NIRS with EEG available in a flexible, adaptable headset for high density mapping with complete mobility with an unlimited range of acquisition for up to 8 hours. MedelOpt Mobility adjusts to real-world situations or virtual reality (VR) environments with a specific design to fit any head mounted VR display. Use for […]View All
Discovering and understanding what motivates humans to produce better results has intrigued many researchers. Many researchers have explored the psychophysiological processes that drive our behavior. Here are some recent studies that have used BIOPAC systems to research motivation… Motivation and Pleasure Deficits Undermine the Benefits of Social Affiliation in Psychosis. Blanchard, J. J., Smith, J. […]
Eye tracking technology has come a long way and has enabled researchers to conduct mobile experiments and track participants in real world scenarios. These featured studies demonstrate some of the use cases for mobile eye tracking technology. Here are some recent studies that have used BIOPAC systems for eye tracking research… Drivers’ gaze patterns when resuming […]Read All