The Nerve Chamber can be connected to biopotential amplifiers and/or a stimulator.
Biopotential amplifiers (MCE100C, ERS100C, or EMG100C) connect via
three JUMP100 connectors and three CBL200 adapter cables
or, for additional lead length, one MEC110C extension cable
Stimulator (STM100C) connects via
one CBL106 and one CBL102
Important—If the STM100C Stimulator is used with a biopotential amplifier on the same nerve, which is nearly always the case, make sure that the black lead of the CBL106 (stimulation negative) is connected to the same pin as the ground lead going to the biopotential amplifier. This is easy to do because the design of the JUMP100 allows stacking connections.
For further details, check the MP Hardware Guide, which is also available under the AcqKnowledge HELP menu.
The ERS Evoked Response Smart Amplifier is an extremely low noise differential amplifier that accurately amplifies very small potentials, such as those associated with stimulus/response, startle response, ABR auditory brainstem response, somatosensory evoked response, signal-averaged recordings, and nerve conduction velocity recordings. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by […]
The mind and body work together in many ways and psychophysiology research provides insight to these connections. BIOPAC has a variety of systems to power your emotion, arousal, stress, cognitive assessment, group interaction, startle, and other psychophysiology research. Here are some recent studies that have used BIOPAC systems for psychophysiological research… Skin conductance, facial EMG, […]
For immediate release: January 4, 2023 Amiens, France and GOLETA, California, USA Seenel Imaging SAS and BIOPAC Systems, Inc. have formed a long-term global partnership to distribute MedelOpt®, Seenel Imaging’s groundbreaking wearable neuroimaging device for the neuroscience research market. Seenel Imaging, based in France, continues its international expansion by partnering with BIOPAC Systems, […]