The Nerve Chamber can be connected to biopotential amplifiers and/or a stimulator.
Biopotential amplifiers (MCE100C, ERS100C, or EMG100C) connect via
three JUMP100 connectors and three CBL200 adapter cables
or, for additional lead length, one MEC110C extension cable
Stimulator (STM100C) connects via
one CBL106 and one CBL102
Important—If the STM100C Stimulator is used with a biopotential amplifier on the same nerve, which is nearly always the case, make sure that the black lead of the CBL106 (stimulation negative) is connected to the same pin as the ground lead going to the biopotential amplifier. This is easy to do because the design of the JUMP100 allows stacking connections.
For further details, check the MP Hardware Guide, which is also available under the AcqKnowledge HELP menu.
The fEMG Electromyography Smart Amplifier is designed specifically for recording facial EMG signals and electrical activity from other small muscle groups. Use with AcqKnowledge™ software to analyze facial expressions and startle paradigms. BIOPAC’s new Smart Amplifiers are designed for great data. Smart Amplifiers improve performance by amplifying the physiological signal close to the subject, which allows a high-level […]
Recent psychophysiological studies feature BIOPAC’s MP Research series of Data Recorders and AcqKnowledge software. Creating Adaptive Assistants for helicopter crews based on Real-time Physiological Data In the Mercedes S-Class, biometric tools track a driver’s eyes, heart rate, and other physiological data, with one goal in mind: Making sure drivers are alert and safe on the […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]