The Nerve Chamber can be connected to biopotential amplifiers and/or a stimulator.
Biopotential amplifiers (MCE100C, ERS100C, or EMG100C) connect via
three JUMP100 connectors and three CBL200 adapter cables
or, for additional lead length, one MEC110C extension cable
Stimulator (STM100C) connects via
one CBL106 and one CBL102
Important—If the STM100C Stimulator is used with a biopotential amplifier on the same nerve, which is nearly always the case, make sure that the black lead of the CBL106 (stimulation negative) is connected to the same pin as the ground lead going to the biopotential amplifier. This is easy to do because the design of the JUMP100 allows stacking connections.
For further details, check the MP Hardware Guide, which is also available under the AcqKnowledge HELP menu.
The NICO Smart Amplifier is specifically designed to measure thoracic impedance for performing noninvasive cardiac output measurements. The amplifier terminates in two connectors for impedance (Z) and its derivative (dZ), and has a 3 m cable that connects directly to an AMI100D Amplifier Input Module. Use with EL526 strip electrodes and CBL246 adapter or with EL500 […]
The autonomic nervous system (ANS) is responsible for regulating the functions of all the body’s organs. Take a look at this month’s citations to see how to apply BIOPAC’s tools and learn techniques for ANS research… Comparable responses to a wide range of olfactory stimulation in women and men. Scientific Reports, 13(1), 9059. Lillqvist, M., Claeson, […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]