The Nerve Chamber can be connected to biopotential amplifiers and/or a stimulator.
Biopotential amplifiers (MCE100C, ERS100C, or EMG100C) connect via
three JUMP100 connectors and three CBL200 adapter cables
or, for additional lead length, one MEC110C extension cable
Stimulator (STM100C) connects via
one CBL106 and one CBL102
Important—If the STM100C Stimulator is used with a biopotential amplifier on the same nerve, which is nearly always the case, make sure that the black lead of the CBL106 (stimulation negative) is connected to the same pin as the ground lead going to the biopotential amplifier. This is easy to do because the design of the JUMP100 allows stacking connections.
For further details, check the MP Hardware Guide, which is also available under the AcqKnowledge HELP menu.
Eyes are the window to the soul and, more importantly, the brain. Psychophysiology, neuroscience, and consumer neuroscience researchers use eye tracking technology to understand emotion, behavior, subject response, decision-making, and human performance—and to help improve products and services. But not everyone knows where to start or how and why to use this important technology. A panel of eye tracking experts will present typical use cases and the latest eye tracking technology.
Join this online presentation with Q&A to learn the fundamentals of eye tracking!
Recent physiological studies feature BIOPAC’s Wireless BioNomadix Data Recorders Using Metronomes and Rhythmic stimulation to help people with Parkinson’s disease walk For those living with Parkinson’s disease, even simple tasks can take too long to accomplish properly, such as crossing a room or picking up an object. In this study, participants with diagnosed cases of […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]