For most biopotential recordings, it is important to provide a low impedance contact with the tissue underneath the skin at the electrode site. Consequently skin should be abraded with pumice, an ELPAD or ELPREP. Alcohol prep pads are generally not preferred because they dry the skin. The purpose of abrasion is to remove dead cells from the outer layer of skin. These cells form an electrical barrier. Electrical communication from the electrode surface to the underlying tissue is most often through a water-based gel. For good recordings, this gel must soak into the skin. As alcohol dries out the skin, it slows the process of establishing the liquid bridge between electrode and tissue.
One exception where abrasion is not recommended is measurement of electrodermal activity (skin conductance). Here skin should be rinsed but neither washed with soap nor abraded. As the conductivity of the skin at the electrode site is the variable to be measured in this case, no steps should be taken to modify that conductance artificially.
Offering the industry’s only system that delivers a comprehensive, 24-hour data logging solution in an easy-to-use, easy-to-wear package, the BioNomadix Logger® truly delivers “physiology anywhere.” The BioNomadix Logger is the perfect tool for applications that demand greater degrees of subject freedom and complex experimental design. BioNomadix Loggers wirelessly record physiological data as subjects freely and naturally […]
BIOPAC provides software and hardware that allows engineers to study the body. Here are a few notable studies using BIOPAC equipment for Biomedical Engineering research. Rehabilitating the Rehab Industry Could physical therapy be done from the comforts of your own home? If possible, this would open the door for more people with all sorts of ailments to […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]