The following is a brief overview of the steps to verify that the USBETH is installed correctly—for more detailed instructions, click to open App Note 280 – ETHUSB Not Communicating on Mac.
1. Download and install at least version 4.0.0 of the driver for the adapter from this website. The driver should be appropriate for the Mac operating system being used.
2. After computer reboots, verify that the adapter is recognized.
- Open System Profiler. Connect ETHUSB. In USB tab, verify that the adapter appears:
3. Go to Software > Extensions and search for AX88772 in the list and verify that version 4.0.0 is installed:
4. Configure the network. Open System Preferences > Network. Select “USB Ethernet” to adjust the adapter settings. Set them to the following manual settings:
- Configure: Manually
- IP: 169.254.142.1
- Subnet Mask: 255.255.0.0
5. Optional: Adjust the network ordering such that USB Ethernet is first on the list.
a) Go to the “gear” at the bottom of the list and choose “Set Service Order” from the list.
b) In the dialog, drag “USB Ethernet” above Ethernet in the list.
c) Click OK and close System Preferences to apply the network settings.
6. Turn on the MP150 (or power cycle it if it was on).
7. Check the amber light on the ETHUSB.
- The amber light should turn on when the MP150 is on and may blink periodically but should generally be solid amber.
8. Launch AcqKnowledge.
- It should now be possible to establish communications with the MP150.
For Mac OS X 10.8 or above
- In section (2), the driver version will be listed as 1.0.0 if the 10.9 driver is installed.
- Add the following intermediate steps:
a) (2.5) Now the USB network interface must be added (10.9+). Open System Preferences > Network. Click on the “+” in the lower left to “Create a new service.
b) In the next dialog that appears, select the “USB Ethernet” interface and give it a default service name. “USB Ethernet” will now be available in the list in the main dialog.
c) (3.5) Unlike previous operating systems, when the cable is connected it will show up as functional with an MP150 when it has a yellow dot.
Electrodermal Activity (EDA) is an objective index for understanding emotional states. Researchers measure EDA to gain insights on psychological or physiological response, leading them down the path of scientific discovery. Because EDA is so important to research, BIOPAC has created tools to optimize event-related EDA analysis. Topics include: how to identify skin conductance responses, convert stim events from E-Prime SuperLab, Presentation, Vizard, etc., into AcqKnowledge events that can used for analysis, and score skin conductance responses using the AcqKnowledge automated EDA analysis tool. Register Now
NIRS and CNAP—Pairing great tools for great data Cerebral hemoglobin concentration (blood mass in the brain) is frequently measured in the prefrontal cortex, and systemic arterial blood pressure is known to be a variable factor based on the hemoglobin concentration. A leg pressure cuff was used to constrict and release blood flow to create a […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […] Read All