The EOG100C amplifier can record from DC to 100 Hz and is designed to measure a signal correlated with the corneal-retinal potential (the small electrical potential difference between the cornea and the retina which probably results from the higher metabolic rate of retinal tissue). EOG is useful for recording eye movements. With electrodes placed around the eyes, the recorded signal will vary with eye position as the cornea moves closer to one electrode and farther from the other (as the retina does the opposite). The corneal-retinal potential does not vary with eye position, but the extent to which the electrodes experience the effects of that potential vary with eye position.
The EMG100C amplifier can record from 1 Hz to 5000 Hz and is designed to measure muscle biopotential signals. It is intended for higher frequency measurements and cannot be used to record DC potentials.
Although the amplifiers are designed to pick up different signals, the EOG100C can pick up some EMG activity (whatever part of the signal is under 100 Hz) and the EMG100C can pick up some EOG activity (but the acquired signal will be different because the unit cannot record the lowest frequencies that the EOG100C records even when the latter is high pass filtered).
Page last modified 24Dec2014
Basic principles of physiological data collection covers fundamentals of data collection and equipment configuration equipment to help you refine experiment protocols and avoid costly missteps. Frazer Findlay, CEO of BIOPAC, discusses common mistakes with physiology recording and shares his secrets for collecting great data. Topics include: How to prep a subject and where to place electrodes; Which type of electrodes work for different body signals; Analog to Digital Conversion; Sample rates (correct vs. incorrect); Filtering; and Scaling and calibration.Watch On Demand Now! Register Now
Recent psychophysiological studies feature BIOPAC’s MP Research series of Data Recorders and AcqKnowledge software. Creating Adaptive Assistants for helicopter crews based on Real-time Physiological Data In the Mercedes S-Class, biometric tools track a driver’s eyes, heart rate, and other physiological data, with one goal in mind: Making sure drivers are alert and safe on the […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […] Read All