Circuit Detail: The “Electrode Check” circuit injects a low current 30 Hz sine wave into the “+” and “-” connectors of the SS1LA or SS2L leads.
The circuit is able to measure the Voltage Peak to Peak (V P-P) between “+” and the Ground electrode, as well as between “-” and the Ground electrode. The voltage will vary depending on the resistance between the electrodes. The more resistance, the more the voltage measured.
The circuit is also able to temporarily remove the input connections, in order to determine the voltage in a high impedance state. This allows an auto calibration to be performed without requiring the user to remove the cable from the “Electrode Check” connector.
The calculated resistance (calculated by the hardware/software/firmware) is displayed on the screen when you choose MP3X menu > Electrode Checker.
If the “Check Electrode Impedance” button is not displayed in BSL EEG Lessons (L03 and L04), you will need to update to restore functionality and prevent potential hardware damage.
Under severe cases of electrostatic discharge, the Electrode Check on older MP36 units (2012 or earlier) or MP35 units could potentially lead to hardware damage. The system will withstand extreme amounts of static discharge and exceeds all of the current standards. To prevent potential damage to these older units, EEG lessons were updated to disable the “Check Impedance” button in BSL EEG Lessons.
Learn a technique—Trans-radial Electrical Bioimpedance Velocimetry (TREV)—that is more user-friendly and less invasive than the transthoracic method. TREV measures blood movement in the radial and ulnar arteries of the forearm via bioimpedance.
– Presentation and discussion of TREV
– Derived Indices Relevant to TREV
– Hardware setup
– Participant setup
– Recognizing good quality and bad quality data
– Demonstration with the grip anticipation test
– Review of results from other tests
Watch On Demand
Exercise physiology is the study of the body’s response to physical activity. Research in exercise physiology allows for insights in health and for a better understanding of the functions of the body at work. Here are some recent studies in exercise physiology that utilized BIOPAC tools for their research… Effect of exercise on electrocardiographic parameters […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]Read All