The “Electrode Check” port on the front panel of MP36 and MP35 units is dseigned to measure impedance.
As a rule of thumb, impedance levels below 10 kOhms are ideal, and the lower the better.
Circuit Detail: The “Electrode Check” circuit injects a low current 30 Hz sine wave into the “+” and “-” connectors of the SS1LA or SS2L leads.
The circuit is able to measure the Voltage Peak to Peak (V P-P) between “+” and the Ground electrode, as well as between “-” and the Ground electrode. The voltage will vary depending on the resistance between the electrodes. The more resistance, the more the voltage measured.
The circuit is also able to temporarily remove the input connections, in order to determine the voltage in a high impedance state. This allows an auto calibration to be performed without requiring the user to remove the cable from the “Electrode Check” connector.
The calculated resistance (calculated by the hardware/software/firmware) is displayed on the screen when you choose MP3X menu > Electrode Checker.
If the “Check Electrode Impedance” button is not displayed in BSL EEG Lessons (L03 and L04), you will need to update to restore functionality and prevent potential hardware damage.
Under severe cases of electrostatic discharge, the Electrode Check on older MP36 units (2012 or earlier) or MP35 units could potentially lead to hardware damage. The system will withstand extreme amounts of static discharge and exceeds all of the current standards. To prevent potential damage to these older units, EEG lessons were updated to disable the “Check Impedance” button in BSL EEG Lessons.
If you have an older MP36: BIOPAC is offering a free Electrode Check hardware upgrade to enhance the performance of the system. After the upgrade, the Electrode Check will be enabled in EEG Lessons 3 and 4. We strongly recommend this upgrade if you use the Electrode Check feature. Please contact Support to register for the free Electrode Check Upgrade.
If you have an older MP35: the Electrode Check can’t be upgraded so we suggest you upgrade to the current MP36 System (MP36U-W/M).
The ETVision (ETV) wearable unit resembles an eyeglasses frame that can be worn by itself or over the participant’s prescription eyeglasses. It contains miniature “eye cameras” that view each eye and a “scene camera” that views the scene in front of the participant, and a microphone. An optional visor can be mounted to the front […]
BIOPAC provides software and hardware that allows research teams to study stimulus presentation data. Here are a few notable studies using BIOPAC equipment for research on stimuli’s effects using ECG and SCL data. Your Face Scares Me This study addressed the impact of individual differences in social anxiety by examining the effects of perceptual load and stimulus valence when […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]