The “Electrode Check” port on the front panel of MP36 and MP35 units is dseigned to measure impedance.
As a rule of thumb, impedance levels below 10 kOhms are ideal, and the lower the better.
Circuit Detail: The “Electrode Check” circuit injects a low current 30 Hz sine wave into the “+” and “-” connectors of the SS1LA or SS2L leads.
The circuit is able to measure the Voltage Peak to Peak (V P-P) between “+” and the Ground electrode, as well as between “-” and the Ground electrode. The voltage will vary depending on the resistance between the electrodes. The more resistance, the more the voltage measured.
The circuit is also able to temporarily remove the input connections, in order to determine the voltage in a high impedance state. This allows an auto calibration to be performed without requiring the user to remove the cable from the “Electrode Check” connector.
The calculated resistance (calculated by the hardware/software/firmware) is displayed on the screen when you choose MP3X menu > Electrode Checker.
If the “Check Electrode Impedance” button is not displayed in BSL EEG Lessons (L03 and L04), you will need to update to restore functionality and prevent potential hardware damage.
Under severe cases of electrostatic discharge, the Electrode Check on older MP36 units (2012 or earlier) or MP35 units could potentially lead to hardware damage. The system will withstand extreme amounts of static discharge and exceeds all of the current standards. To prevent potential damage to these older units, EEG lessons were updated to disable the “Check Impedance” button in BSL EEG Lessons.
If you have an older MP36: BIOPAC is offering a free Electrode Check hardware upgrade to enhance the performance of the system. After the upgrade, the Electrode Check will be enabled in EEG Lessons 3 and 4. We strongly recommend this upgrade if you use the Electrode Check feature. Please contact Support to register for the free Electrode Check Upgrade.
If you have an older MP35: the Electrode Check can’t be upgraded so we suggest you upgrade to the current MP36 System (MP36U-W/M).
High-performance mobile fNIR Imaging Systems are stand-alone functional brain imaging solutions for continuous NIR spectroscopy (NIRS). These top of the line systems provide cognitive function assessment and eliminate a great many of the drawbacks of a functional MRI. Mobile subjects are comfortable and can respond naturally in real-world situations to take a test or perform mobile […]
BIOPAC provides software and hardware that allows research teams to run autonomic function tests while collecting physiological data. Here are a few studies focusing on BIOPAC’s ability to record physiological signals during autonomic function testing. The Internet, Sleep, and Heart Rate Does internet usage lead to poorer sleep? Using a BIOPAC Data Acquisition System to measure cardiovascular […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]