The BSL MP30/35/36 unit has four isolated, human-safe universal biopotential/transducer amplifiers that are controlled by the BSLPRO software. When recording data, it is important to set the gain of the amplifier as high as possible without clipping it. If you are unsure what gain to use for your signal type, the OptimizeRanges feature will adjust the screen display so that you can see the incoming signal relative to the range of the A/D converter. The OptimizeRanges option is located from the Display menu, by right mouse clicking, and also as a toolbar icon . A green bar located at the right edge of the channel display identifies the vertical range.
If the signal is very small, relative to the display window, increase the gain so that the signal fills more of the channel display region. This will maximize the resolution of the signal relative to the input signal range of the A/D converter. The following screen shots show an ECG signal with the gain set to 1K, 5K, and 10K. By increasing the gain of the amplifier to 10K, the signal is using the full range of the A/D converter.
Note: The OptimizeRanges feature only works on analog input channels.
The following screen shows data with the default ECG gain setting (1K).
The following screen shows the data with the gain set to 5K.
The following screen shows the data with the gain set to 10K.
The gain is adjusted in the Setup channels > View change parameters dialog
Smart Center Stand-alone, Portable Wireless Physiology Systems Smart Center Essentials entry-level system includes a Smart Center Device (BN-SMART with USB power cable), 2 or 3 Transmitters (BN-xxx-T with chargers); AcqKnowledge for Smart Center, and a case. Smart Center Enhanced adds a Logger (BN-LOGGER) and Basic Scripting License (ACK100W-BAS) to the Smart Center Device (BN-SMART with powercable), 3 […]
Recent physiological studies feature BIOPAC’s Wireless BioNomadix Data Recorders Using Metronomes and Rhythmic stimulation to help people with Parkinson’s disease walk For those living with Parkinson’s disease, even simple tasks can take too long to accomplish properly, such as crossing a room or picking up an object. In this study, participants with diagnosed cases of […]
BIOPAC’s just released Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines […]