The BSL MP30/35/36 unit has four isolated, human-safe universal biopotential/transducer amplifiers that are controlled by the BSLPRO software. When recording data, it is important to set the gain of the amplifier as high as possible without clipping it. If you are unsure what gain to use for your signal type, the OptimizeRanges feature will adjust the screen display so that you can see the incoming signal relative to the range of the A/D converter. The OptimizeRanges option is located from the Display menu, by right mouse clicking, and also as a toolbar icon . A green bar located at the right edge of the channel display identifies the vertical range.
If the signal is very small, relative to the display window, increase the gain so that the signal fills more of the channel display region. This will maximize the resolution of the signal relative to the input signal range of the A/D converter. The following screen shots show an ECG signal with the gain set to 1K, 5K, and 10K. By increasing the gain of the amplifier to 10K, the signal is using the full range of the A/D converter.
Note: The OptimizeRanges feature only works on analog input channels.
The following screen shows data with the default ECG gain setting (1K).
The following screen shows the data with the gain set to 5K.
The following screen shows the data with the gain set to 10K.
The gain is adjusted in the Setup channels > View change parameters dialog
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
Comparing Diaphragm Tissue Between Healthy and Ill Patients The human diaphragm is one of the primary muscles used in respiration, contracting and expanding to control breathing. To measure how much diaphragms move for critically ill patients (in an Intensive Care Unit) compared to diaphragms in a healthy subject, patients at “Papageorgiou” General Hospital in Thessaloniki, […]
BIOPAC’s ventilator testing equipment and expertise supports STARK Industries’ effort to manufacture, validate, and test NASA JPL’s VITAL Ventilators Goleta, California – June 16, 2020 BIOPAC Systems, Inc. announced a partnership with STARK Industries, LLC to provide ventilator validation and testing equipment for the mass production of ventilators. STARK Industries was awarded a global license […]