The BSL MP30/35/36 unit has four isolated, human-safe universal biopotential/transducer amplifiers that are controlled by the BSLPRO software. When recording data, it is important to set the gain of the amplifier as high as possible without clipping it. If you are unsure what gain to use for your signal type, the OptimizeRanges feature will adjust the screen display so that you can see the incoming signal relative to the range of the A/D converter. The OptimizeRanges option is located from the Display menu, by right mouse clicking, and also as a toolbar icon . A green bar located at the right edge of the channel display identifies the vertical range.
If the signal is very small, relative to the display window, increase the gain so that the signal fills more of the channel display region. This will maximize the resolution of the signal relative to the input signal range of the A/D converter. The following screen shots show an ECG signal with the gain set to 1K, 5K, and 10K. By increasing the gain of the amplifier to 10K, the signal is using the full range of the A/D converter.
Note: The OptimizeRanges feature only works on analog input channels.
The following screen shows data with the default ECG gain setting (1K).
The following screen shows the data with the gain set to 5K.
The following screen shows the data with the gain set to 10K.
The gain is adjusted in the Setup channels > View change parameters dialog
BSL Home enables students to safely conduct hands-on physiology labs wherever they are! Purchase this complete lab solution to preserve practical lab experience! Includes MP41 Unit+Lead Set with BSL Home software, EL503 electrodes, WRAP1 self-adhering wrap for EEG lessons, and disposable 9 V battery. For instructors—BSL Home makes it easy for instructors to continue running practical […]
BIOPAC provides software and hardware that allows research teams to record and analyze numerous body signals for physiological experimentation. Here are a few notable studies using BIOPAC equipment for ECG, EDA, and Respiration data. What Keeps Students More Engaged? The question has been brought to the forefront during COVID, what variations of online learning keep students more engaged? […]
BIOPAC’s comprehensive Introductory ECG Guide addresses fundamental to advanced concerns to optimize electrocardiography data recording and analysis. Topics include: ECG Complex; Electrical and Mechanical Sequence of a Heartbeat; Systole and Diastole; Configurations for Lead I, Lead II, Lead III, 6-lead ECG, 12-lead ECG, precordial leads; Ventricular Late Potentials (VLPs); ECG Measurement Tools; Automated Analysis Routines for extracting, […]